title: Ultra-pressure liquid chromatography-electrospray tandem mass spectrometry for multiresidue determination of pesticides in water.
abstract: A multiresidue analysis method has been developed for the determination of pesticides in water by ultra-performance liquid chromatography (UPLC) combined with tandem mass spectrometry (MS/MS). The selected pesticides represent a broad range of polarity and volatility [benzoylcyclohexanedione (mesotrione and sulcotrione); chloroacetamide (acetochlor, alachlor, dimethenamide, and metolachlor); phenoxyacetic acid (2,4-D and MCPA); phenoxypropionic (dichloprop and mecoprop); phenylurea (chlortoluron, diuron, isoproturon, linuron, and metoxuron); sulfonylurea (foramsulfuron, iodosulfuron, and nicolsulfuron); triazine (atrazine, cyanazine, desethylatrazine (DEA), desisopropylatrazine (DIA), simazine, and terbutylazine)]. The analytes were extracted using solid-phase extraction (SPE). The separation was carried out on an acquity UPLC BEH C18 column (1.7 microm, 50 mm x 1 mm ID) using a gradient elution profile and mobile phase consisting of 0.1% formic acid in water and acetonitrile. The pesticides were detected with a tandem mass spectrometer after being ionised positively or negatively (depending on the molecule) using an electrospray ionisation (ESI) source. To achieve the suitable extraction conditions for sample preparation, several parameters affecting the efficiency of SPE such as the nature of the sorbent and the eluent, extractant volume and pH were studied. The best recovery was obtained by the extraction with an Oasis HLB cartridge and 3 mL of a solution of acetonitrile/dichloromethane (1:1, v/v) at pH 2. The average recoveries of the pesticides in different samples ranged from 82 to 109%. The weight least squares (WLS) linear regression was used to calculate the limits of detection and quantification (LOD and LOQ) because the dispersion was heteroskedastic. All the pesticides could be correctly quantified at a concentration level of 50 ng L(-1) and most of them could be detected at a concentration inferior or equal to 8 ng L(-1). Efficiency and robustness of this method were evaluated by the analysis of several samples of real natural water.
authors.affiliation: Laboratoire Sciences Chimiques de Rennes UMR CNRS 6226/ENSCR/UR1, Equipe CIP, Ecole Nationale Supérieure de Chimie de Rennes, avenue du Général Leclerc, 35700 Rennes, France.
authors.forename: G
authors.initials: G
authors.lastname: Gervais
authors.forename: S
authors.initials: S
authors.lastname: Brosillon
authors.forename: A
authors.initials: A
authors.lastname: Laplanche
authors.forename: C
authors.initials: C
authors.lastname: Helen
mesh_terms: D000097:Acetonitriles / Q000737:chemistry; D002851:Chromatography, High Pressure Liquid / Q000379:methods*; D006863:Hydrogen-Ion Concentration; D008752:Methylene Chloride / Q000737:chemistry; D010573:Pesticide Residues / Q000032:analysis / Q000302:isolation & purification; D010575:Pesticides / Q000032:analysis* / Q000302:isolation & purification; D015203:Reproducibility of Results; D052616:Solid Phase Extraction; D021241:Spectrometry, Mass, Electrospray Ionization / Q000379:methods*; D053719:Tandem Mass Spectrometry / Q000379:methods*; D014874:Water Pollutants, Chemical / Q000032:analysis* / Q000302:isolation & purification
chemical_list: D000097:Acetonitriles; D010573:Pesticide Residues; D010575:Pesticides; D014874:Water Pollutants, Chemical; D008752:Methylene Chloride; C032159:acetonitrile