Real-time method for NO(2) flow measurements.
Measurements of NO(2) flows are made without requiring prior experimental calibration of the flow meter. A tapered, variable area flow meter with a diameter ratio scale is used to make the real-time NO(2) flow measurements. The necessary parameters and calculations used to determine the viscosity and density of the flowing N(2)O(4)2NO(2) gas are presented. The limits of error associated with these parameters and errors associated with controlled flow measurement conditions are considered. Disregarding the scale reading error, which depends on the flow meter utilized, the limits of error are found to yield a measurement error less than 7%.
Darrah
R C
RC
Universal Energy Systems, Dayton, Ohio 45432Southeastern Center for Electrical Engineering Education, Auburn, Alabama 36830Air Force Aero Propulsion Lab., Wright-Patterson AFB, Dayton, Ohio 45432.
Andrews
M L
ML
Garscadden
A
A
Bletzinger
P
P