100-MHz noise spectrometer.
The design, construction, and performance of a 100-MHz noise spectrometer are described. Basically a 100-MHz bridge, this system uses the properties of a double balanced mixer to enable the experimenter to determine spectral densities of fluctuations in the real or imaginary parts of a sample impedance, as well as the real part of the cross-spectral density. Calibration circuits and techniques are described. The results of measurements on commercial 50-Omega terminators demonstrate fluctuations of both the real and the imaginary parts of the impedance and partial correlation between them.
Rzemien
R
R
Department of Physics and Astrophysics, University of Colorado, Boulder, Colorado 80309.
Iddings
C K
CK
Love
W F
WF