title: [In vitro evaluation of low-temperature aging effects of Y2O3 stabilized tetragonal zirconia polycrystals dental ceramics].
abstract: OBJECTIVE
  To investigate the influence of in vitro low-temperature degradation (LTD) treatment on the structural stability of 5 kinds of Y2O3 stabilized tetragonal zirconia polycrystals (Y-TZP) dental ceramics.
  
  
  METHODS
  TZ-3YS powder was compacted at 200 MPa using cold isostatic pressure and pre-sintered at 1050 degrees C for 2 h forming presintered blocks. Specimens were sectioned into 15 mm x 15 mm x 1.5 mm slices from blocks of TZ-3YS, Vita In-Ceram YZ, Ivoclar, Cercon Smart, and Kavo Y-TZP presintered blocks, 18 slices for each brand, and then densely sintered. Specimens were divided into 6 groups and subjected to an accelerated aging test carried out in an autoclave in steam at 134 degrees C, 0.2 MPa, for 0, 1, 2, 3, 4, and 5 h. X-ray diffraction (XRD) was used to identify crystal phases and relative content of monoclinic phase was calculated. Specimens for three-point bending test were fabricated using TZ-3YS ceramics according to the ISO 6872 standard and bending strength was tested before and after aging. The polished and aging specimens of TZ-3YS and Cercon Smart zirconia ceramics were observed by atomic force microscopy (AFM) to evaluate surface microstructure.
  
  
  RESULTS
  Tetragonal-to-monoclinic phase transformation was detected for specimens of TZ-3YS, Vita In-Ceram YZ, Ivoclar, and Kavo zirconia ceramics except for Cercon Smart ceramics after aging, and the relative content of monoclinic phase was increasing with the prolonged aging time. TZ-3YS was the most affected material, Kavo took the second, and Vita and Ivoclar were similar. Aging had no significant negative effects on flexural strength of TZ-3YS with average bending strength being over 1100 MPa. The nucleation and growth of monoclinic phase were detected by AFM in surface of Cercon Smart zirconia in which monoclinic phase was not detected by XRD.
  
  
  CONCLUSIONS
  The results suggest that LTD of dental Y-TZP is time dependent, but the aging test does not reduce the flexural strength of TZ-3YS. The long-term clinical serviceability of dental Y-TZP needs further observation.
authors.affiliation: Department of Stomatology, General Hospital of PLA, Beijing, China.
authors.forename: Yuan-fu
authors.initials: YF
authors.lastname: Yi
authors.forename: Hong-chen
authors.initials: HC
authors.lastname: Liu
authors.forename: Chen
authors.initials: C
authors.lastname: Wang
authors.forename: Jie-mo
authors.initials: JM
authors.lastname: Tian
authors.forename: Ning
authors.initials: N
authors.lastname: Wen
mesh_terms: D003080:Cold Temperature; D003776:Dental Porcelain / Q000737:chemistry*; D003799:Dental Stress Analysis; D008422:Materials Testing; D013997:Time Factors; D015040:Zirconium / Q000737:chemistry*
chemical_list: D003776:Dental Porcelain; D015040:Zirconium; C028541:zirconium oxide