Micro insert: a prototype full-ring PET device for improving the image resolution of a small-animal PET scanner.
A full-ring PET insert device should be able to enhance the image resolution of existing small-animal PET scanners.


METHODS
The device consists of 18 high-resolution PET detectors in a cylindric enclosure. Each detector contains a cerium-doped lutetium oxyorthosilicate array (12 x 12 crystals, 0.72 x 1.51 x 3.75 mm each) coupled to a position-sensitive photomultiplier tube via an optical fiber bundle made of 8 x 16 square multiclad fibers. Signals from the insert detectors are connected to the scanner through the electronics of the disabled first ring of detectors, which permits coincidence detection between the 2 systems. Energy resolution of a detector was measured using a (68)Ge point source, and a calibrated (68)Ge point source stepped across the axial field of view (FOV) provided the sensitivity profile of the system. A (22)Na point source imaged at different offsets from the center characterized the in-plane resolution of the insert system. Imaging was then performed with a Derenzo phantom filled with 19.5 MBq of (18)F-fluoride and imaged for 2 h; a 24.3-g mouse injected with 129.5 MBq of (18)F-fluoride and imaged in 5 bed positions at 3.5 h after injection; and a 22.8-g mouse injected with 14.3 MBq of (18)F-FDG and imaged for 2 h with electrocardiogram gating.


RESULTS
The energy resolution of a typical detector module at 511 keV is 19.0% +/- 3.1%. The peak sensitivity of the system is approximately 2.67%. The image resolution of the system ranges from 1.0- to 1.8-mm full width at half maximum near the center of the FOV, depending on the type of coincidence events used for image reconstruction. Derenzo phantom and mouse bone images showed significant improvement in transaxial image resolution using the insert device. Mouse heart images demonstrated the gated imaging capability of the device.


CONCLUSIONS
We have built a prototype full-ring insert device for a small-animal PET scanner to provide higher-resolution PET images within a reduced imaging FOV. Development of additional correction techniques are needed to achieve quantitative imaging with such an insert.
Wu
Heyu
H
Mallinckrodt Institute of Radiology, Washington University, St. Louis, Missouri, USA.
Pal
Debashish
D
Song
Tae Yong
TY
O'Sullivan
Joseph A
JA
Tai
Yuan-Chuan
YC
D000465:Algorithms; D000818:Animals; D003952:Diagnostic Imaging / Q000295:instrumentation / Q000379:methods; D004562:Electrocardiography / Q000379:methods; D004867:Equipment Design; D005462:Fluorine Radioisotopes / Q000494:pharmacology; D019788:Fluorodeoxyglucose F18 / Q000494:pharmacology; D007091:Image Processing, Computer-Assisted; D051379:Mice; D009683:Nuclear Medicine / Q000379:methods; D019047:Phantoms, Imaging; D049268:Positron-Emission Tomography / Q000295:instrumentation* / Q000379:methods*
D005462:Fluorine Radioisotopes; D019788:Fluorodeoxyglucose F18
Ann Med. 1999 Dec;31(6):432-7
10680858
Phys Med Biol. 1995 Jun;40(6):1105-26
7659733
J Nucl Med. 2008 Jan;49(1):79-87
18077526
J Nucl Med. 2005 Mar;46(3):455-63
15750159
J Nucl Med. 2000 Apr;41(4):661-81
10768568
J Nucl Med. 2008 Mar;49(3):471-9
18287272
Mol Imaging. 2005 Apr-Jun;4(2):117-27
16105510
Phys Med Biol. 2005 Jan 21;50(2):179-95
15742938
Annu Rev Biomed Eng. 2005;7:255-85
16004572
Phys Med Biol. 2003 Jun 7;48(11):1519-37
12817935
Radiology. 2001 May;219(2):316-33
11323453
Phys Med Biol. 1999 Mar;44(3):781-99
10211810
J Nucl Med. 1987 Nov;28(11):1717-24
3499493
Phys Med Biol. 2007 Jul 21;52(14):4293-310
17664609