title: Validation of an off-eye contact lens Shack-Hartmann wavefront aberrometer.
abstract: OBJECTIVE
  To evaluate the ability of a commercially available single pass Shack-Hartmann aberrometer to evaluate contact lens aberrations.
  
  
  METHODS
  Accuracy of second-order aberration measurements was verified by measuring a series of precision calibration lenses, spectacle lenses, and contact lenses. Power measurements were compared to those expected by an independent measurement or those provided by the lens manufacturer. Accuracy of third-order aberrations was verified by systematically decentering a lens with known amounts of spherical aberration and comparing the magnitude of induced coma to that of optical theory. Fourth-order aberration accuracy was verified by comparing measured longitudinal spherical aberration values to those expected by ray tracing based on the lens design. Accuracy of lower- and higher-order aberrations was verified for measurements of lenses taken in air and within a saline-filled wet cell. Repeatability was also assessed by comparing repeated measurements of the wet cell and lens in a wet cell, before and after manipulation of that cell.
  
  
  RESULTS
  In all cases, measured values closely matched the expected values, generally exhibiting errors of <1%.
  
  
  CONCLUSIONS
  The instrument demonstrates good accuracy and repeatability in measuring second-, third-, and fourth-order aberrations of contact lenses and provides the industry with an instrument for evaluating the ex vivo optical characteristics of contact lenses.
authors.affiliation: Indiana University, School of Optometry, Bloomington, Indiana 47405, USA. kollbaum@indiana.edu
authors.forename: Pete
authors.initials: P
authors.lastname: Kollbaum
authors.forename: Meredith
authors.initials: M
authors.lastname: Jansen
authors.forename: Larry
authors.initials: L
authors.lastname: Thibos
authors.forename: Arthur
authors.initials: A
authors.lastname: Bradley
mesh_terms: D003261:Contact Lenses; D003941:Diagnostic Techniques, Ophthalmological / Q000295:instrumentation*; D006801:Humans; D055620:Optical Phenomena; D009907:Optometry / Q000295:instrumentation*; D015203:Reproducibility of Results
references.citation: Appl Opt. 2005 Jul 20;44(21):4523-7
references.pmid: 16047902
references.citation: Eye Contact Lens. 2003 Jan;29(1 Suppl):S167-70; discussion S190-1, S192-4
references.pmid: 12772758
references.citation: CLAO J. 1995 Jul;21(3):163-8
references.pmid: 7586474
references.citation: J Opt Soc Am A Opt Image Sci Vis. 2001 May;18(5):1003-15
references.pmid: 11336203
references.citation: Am J Optom Physiol Opt. 1988 Apr;65(4):298-307
references.pmid: 3377065
references.citation: Optom Vis Sci. 1998 Jan;75(1):44-54
references.pmid: 9460786
references.citation: Optom Vis Sci. 2003 Aug;80(8):587-95
references.pmid: 12917578
references.citation: Optom Vis Sci. 2003 Feb;80(2):126-34
references.pmid: 12597327
references.citation: Am J Optom Arch Am Acad Optom. 1961 Aug;38:445-8
references.pmid: 13784617
references.citation: Int Contact Lens Clin. 1999 Nov;26(6):168-173
references.pmid: 11384834
references.citation: Opt Lett. 1979 Jul 1;4(7):224-6
references.pmid: 19687857
references.citation: Ophthalmic Physiol Opt. 2003 Nov;23(6):517-25
references.pmid: 14622355
references.citation: Optom Vis Sci. 1990 Apr;67(4):277-82
references.pmid: 2342790
references.citation: J Refract Surg. 2002 Sep-Oct;18(5):S603-9
references.pmid: 12361166
references.citation: Lancet. 1986 Feb 8;1(8476):307-10
references.pmid: 2868172