Multiwalled carbon nanotubes as solid-phase extraction materials for the gas chromatographic determination of organophosphorus pesticides in waters.
In the present work, a GC method with nitrogen-phosphorus detection (NPD) was developed for the simultaneous determination of eight organophosphorus pesticide (OPP) residues (i.e., ethoprofos, diazinon, chlorpyrifos-methyl, fenitrothion, malathion, chlorpyrifos, fenamiphos, and buprofezin) in water samples. Preconcentration of the water samples was carried out using an SPE procedure with multiwalled carbon nanotubes (MWCNTs) of 10-15 nm od, 2-6 nm id, and 0.1-10 microm length as stationary phase. Extraction parameters, such as the amount of MWCNTs, sample volume, pH, and type and amount of the eluent were optimized. The most favorable conditions were as follows: 40 mg MWCNTs, 800 mL water, pH 6.0, and 20 mL dichloromethane, respectively. The MWCNTs-SPE-GC-NPD method was applied to the determination of these pesticides in real water samples: mineral and ground water as well as run-off water from an agricultural area collected shortly before opening out onto the sea. A recovery study was developed with five consecutive extractions of the three types of water spiked at three concentration levels (n = 15). Mean recovery values were in the range of 75-116% for mineral water (RSD < 6.3%), 67-119% for ground water (RSD < 5.8%), and 57-81% for run-off waters (RSDs < 6.9%), except for fenamiphos (mean recovery values between 40 and 84% for the three types of waters, RSDs < 8.9%). LODs were in the low ng/L level (i.e., levels below the maximum residue limits (MRLs) established by the European Union (EU) legislation for these compounds in waters). The proposed method was also applied to the analysis of six water samples (two of each type: mineral, ground, and run-off waters) in which no residues of the selected pesticides were found. Results show that the MWCNTs used in this work have a high adsorbability of the pesticides under study. The main advantage of the use of these MWCNTs is their low cost when compared with those MWCNTs previously used in the literature and with conventional SPE cartridges.
Ravelo-Pérez
Lidia M
LM
Department of Analytical Chemistry, Nutrition and Food Science, University of La Laguna, Tenerife, Canary Islands, Spain.
Hernández-Borges
Javier
J
Angel Rodríguez-Delgado
Miguel
M