PTV-LV-GC/MS/MS as screening and complementary method to HRMS for the monitoring of dioxin levels in food and feed.
Recent developments in trapping efficiency inside ion trap mass spectrometer permitted to lower instrument detection limit (IDL). An IDL of 200fgmul(-1) injected with a signal-to-noise ratio of 5:1 for tetrachlorodibenzo-p-dioxin (TCDD) was obtained by gas chromatography coupled to a quadrupole ion storage mass spectrometer in tandem mode (GC/MS/MS). Coupling large volume programmable temperature vaporizer (PTV-LV) injection to GC/MS/MS provides an alternative and complementary method to classical splitless-GC injection connected to high-resolution mass spectrometry (splitless-GC/HRMS) method for dioxin monitoring in food and feed. An injection volume of 10mul was found to be the best compromise between the sensitivity requirements and the robustness required for a high throughput method. PTV-LV-GC/MS/MS and Splitless-GC/HRMS were compared by performing analysis on five different matrices such as beef fat, yolk eggs, milk powder, animal feed and serum samples covering a concentration range of two orders of magnitude (i.e. 0.2-25ng WHO-TEQkg(-1)). An analysis of variance (ANOVA) was carried out. Fisher tests pointed out that the method effect for all the 2,3,7,8 congeners was not significant, indicating that the null hypothesis (H(0): mu(1)=mu(2)=...=mu(n)) was not rejected. Moreover, the interaction effects between methods and matrices were not significant for most of the 2,3,7,8 congeners. However, three congeners (2,3,7,8-TCDF; 1,2,3,4,7,8-HxCDD and 1,2,3,4,6,7,8-HpCDD) were characterized by P-values lower than the significance level (alpha=0.05). In toxic equivalence (TEQ), the study showed that no significant bias was observed between the two methods. Consequently, PTV-LV-GC/MS/MS is an attractive technique and can be used as a cost effective complementary method to HRMS for dioxin levels monitoring in food and feed.
Eppe
Gauthier
G
Mass Spectrometry Laboratory-C.A.R.T., University of Liège, Allée de la Chimie 3, B-6C Sart-Tilman B-4000, Belgium.
Focant
Jean-François
JF
Pirard
Catherine
C
Pauw
Edwin De
ED