title: Comparison of the load-sharing characteristics between pedicle-based dynamic and rigid rod devices.
abstract: Recently, numerous types of posterior dynamic stabilization (PDS) devices have been introduced as an alternative to the fusion devices for the surgical treatment of degenerative lumbar spine. It is hypothesized that the use of 'compliant' materials such as Nitinol (Ni-Ti alloy, elastic modulus = 75 GPa) or polyether-etherketone (PEEK, elastic modulus = 3.2 GPa) in PDS can restore stability of the lumbar spine without adverse stress-shielding effects that have often been found with 'rigid' fusion devices made of 'rigid' Ti alloys (elastic modulus = 114 GPa). Previous studies have shown that suitably designed PDS devices made of more compliant material may be able to help retain kinematic behavior of the normal spine with optimal load sharing between the anterior and posterior spinal elements. However, only a few studies on their biomechanical efficacies are available. In this study, we conducted a finite-element (FE) study to investigate changes in load-sharing characteristics of PDS devices. The implanted models were constructed after modifying the previously validated intact model of L3-4 spine. Posterior lumbar fusion with three different types of pedicle screw systems was simulated: a conventional rigid fixation system (Ti6Al4V, Phi = 6.0 mm) and two kinds of PDS devices (one with Nitinol rod with a three-coiled turn manner, Phi = 4.0 mm; the other with PEEK rod with a uniform cylindrical shape, Phi = 6.0 mm). To simulate the load on the lumbar spine in a neutral posture, an axial compressive load (400 N) was applied. Subsequently, the changes in load-sharing characteristics and stresses were investigated. When the compressive load was applied on the implanted models (Nitinol rod, PEEK rod, Ti-alloy rod), the predicted axial compressive loads transmitted through the devices were 141.8 N, 109.8 N and 266.8 N, respectively. Axial forces across the PDS devices (Nitinol rod, PEEK rod) and rigid system (Ti-alloy rod) with facet joints were predicted to take over 41%, 33% and 71% of the applied compression load, respectively. Our results confirmed the hypothesis on the PDS devices by showing the substantial reduction in stress-shielding characteristics. Higher axial load was noted across the anterior structure with the PDS devices, which could slow the degeneration process of bony structures and lower the possibility of implant failure.
authors.affiliation: Korea Orthopedics and Rehabilitation Engineering Center, Incheon 403712, Korea.
authors.forename: Yoon-Ho
authors.initials: YH
authors.lastname: Ahn
authors.forename: Wen-Ming
authors.initials: WM
authors.lastname: Chen
authors.forename: Kwon-Yong
authors.initials: KY
authors.lastname: Lee
authors.forename: Kyung-Woo
authors.initials: KW
authors.lastname: Park
authors.forename: Sung-Jae
authors.initials: SJ
authors.lastname: Lee
mesh_terms: D000497:Alloys / Q000737:chemistry; D001696:Biomechanical Phenomena; D019245:Compressive Strength; D004548:Elasticity; D004867:Equipment Design; D020342:Finite Element Analysis; D006801:Humans; D008159:Lumbar Vertebrae / Q000473:pathology*; D008422:Materials Testing; D008962:Models, Theoretical; D016012:Poisson Distribution; D011312:Pressure; D011474:Prosthesis Design; D011475:Prosthesis Failure; D014025:Titanium / Q000737:chemistry
chemical_list: D000497:Alloys; C013616:nitinol; D014025:Titanium