title: Artificial atlanto-odontoid joint replacement through a transoral approach.
abstract: Resection of the odontoid process and anterior arch of the atlas results in atlantoaxial instability, which if left uncorrected may lead to severe neurological complications. Currently, such atlantoaxial instability is corrected by anterior and/or posterior C1-C2 fusion. However, this results in considerable loss of rotation function of the atlantoaxial complex. From the viewpoint of retaining the rotation function and providing stability, we designed an artificial atlanto-odontoid joint based on anatomical measurements of 50 pairs of dry atlantoaxial specimens by digital calipers and 10 fresh cadaveric specimens by microsurgical techniques. The metal-on-metal titanium alloy joint has an arc-shaped atlas component, and a hollow cylindrical bushing into which fits a rotation axle of an inverted v-shaped axis component and is implanted through a transoral approach. After the joint was implanted onto specimens with anterior decompression, biomechanical tests were performed to compare the stability parameters in the intact state, after decompression, after artificial joint replacement, and after fatigue test. Compared to the intact state, artificial joint replacement resulted in a significant decrease in the range of motion (ROM) and neutral zone (NZ) during flexion, extension, and lateral bending (P < 0.001); however, with regard to axial rotation, there was no significant difference in ROM (P = 0.405), a significant increase in NZ (P = 0.008), and a significant decrease in stiffness (P = 0.003). Compared to the decompressed state, artificial joint replacement resulted in a significantly decreased ROM (P B 0.021) and NZ (P B 0.002) and a significantly increased stiffness (P \ 0.001) in all directions. Following artificial joint replacement, there was no significant difference in ROM (P C 0.719), NZ (P C 0.580), and stiffness (P C 0.602) in all directions before and after the fatigue test. The artificial joint showed no signs of wear and tear after the fatigue test. This artificial atlanto-odontoid joint may be useful in cases of odontoid resection due to malunion or nonunion of odontoid fracture, atraumatic odontoid fracture, irreducible atlas dislocation, posterior atlantoaxial subluxation, or congenital skull base abnormalities.
authors.affiliation: Department of Orthopaedic Surgery, The 2nd Hospital of Xi'an Jiaotong University Medical College, Xi'an, People's Republic of China.
authors.forename: Bin
authors.initials: B
authors.lastname: Lu
authors.forename: Xi Jing
authors.initials: XJ
authors.lastname: He
authors.forename: Chen Guang
authors.initials: CG
authors.lastname: Zhao
authors.forename: Hao Peng
authors.initials: HP
authors.lastname: Li
authors.forename: Dong
authors.initials: D
authors.lastname: Wang
mesh_terms: D000328:Adult; D019643:Arthroplasty, Replacement / Q000379:methods*; D001268:Atlanto-Axial Joint / Q000502:physiology / Q000601:surgery*; D001696:Biomechanical Phenomena; D002102:Cadaver; D001270:Cervical Atlas / Q000502:physiology / Q000601:surgery*; D006801:Humans; D007595:Joint Prosthesis; D008422:Materials Testing; D008875:Middle Aged; D009809:Odontoid Process / Q000502:physiology / Q000601:surgery*; D016059:Range of Motion, Articular