ISCEV Standard for full-field clinical electroretinography (2008 update).
This document, from the International Society for Clinical Electrophysiology of Vision (ISCEV), presents an updated and revised ISCEV Standard for clinical electroretinography (ERG). The parameters for flash stimulation and background adaptation have been tightened, and responses renamed to indicate the flash strength (in cd x s x m(-2)). The ISCEV Standard specifies five responses: (1) Dark-adapted 0.01 ERG (rod response); (2) Dark-adapted 3.0 ERG (combined rod-cone response); (3) Dark-adapted 3.0 oscillatory potentials; (4) Light-adapted 3.0 ERG (cone response); (5) Light-adapted 3.0 flicker (30 Hz flicker). An additional Dark-adapted 10.0 ERG or Dark-adapted 30.0 ERG response is recommended.
Marmor
M F
MF
Department of Ophthalmology, Stanford University School of Medicine, Stanford, CA, USA. marmor@stanford.edu
Fulton
A B
AB
Holder
G E
GE
Miyake
Y
Y
Brigell
M
M
Bach
M
M
D000221:Adaptation, Ocular; D000328:Adult; D000368:Aged; D000369:Aged, 80 and over; D002138:Calibration; D002648:Child; D002675:Child, Preschool; D002985:Clinical Protocols / Q000592:standards; D004566:Electrodes; D004596:Electroretinography / Q000295:instrumentation* / Q000592:standards*; D006801:Humans; D007223:Infant; D010775:Photic Stimulation / Q000379:methods; D012107:Research Design / Q000592:standards; D013223:Statistics as Topic / Q000379:methods; D009626:Terminology as Topic