A comparison of shielding calculation methods for multi-slice computed tomography (CT) systems.
Currently in the UK, shielding calculations for computed tomography (CT) systems are based on the BIR-IPEM (British Institute of Radiology and Institute of Physics in Engineering in Medicine) working group publication from 2000. Concerns have been raised internationally regarding the accuracy of the dose plots on which this method depends and the effect that new scanner technologies may have. Additionally, more recent shielding methods have been proposed by the NCRP (National Council on Radiation Protection) from the USA. Thermoluminescent detectors (TLDs) were placed in three CT scanner rooms at different positions for several weeks before being processed. Patient workload and dose data (DLP: the dose length product and mAs: the tube current-time product) were collected for this period. Individual dose data were available for more than 95% of patients scanned and the remainder were estimated. The patient workload data were used to calculate expected scatter radiation for each TLD location by both the NCRP and BIR-IPEM methods. The results were then compared to the measured scattered radiation. Calculated scattered air kerma and the minimum required lead shielding were found to be frequently overestimated compared to the measured air kerma, on average almost five times the measured scattered air kerma.
Cole
J A
JA
KCARE, Medical Engineering and Physics, Faraday Building, King's College Hospital NHS Foundation Trust, London, UK. jonathan.cole@nhs.net
Platten
D J
DJ
D003198:Computer Simulation; D017076:Computer-Aided Design; D004867:Equipment Design / Q000379:methods; D019544:Equipment Failure Analysis / Q000379:methods*; D008962:Models, Theoretical; D011835:Radiation Protection / Q000295:instrumentation* / Q000379:methods; D015203:Reproducibility of Results; D012680:Sensitivity and Specificity; D014057:Tomography, X-Ray Computed / Q000295:instrumentation*; D006113:United Kingdom