Utility of mass spectrometry for proteome analysis: part I. Conceptual and experimental approaches.
This article is the first in a series of reviews intended as a tutorial providing the inexperienced, as well as the experienced, reader with an overview of principles of peptide and protein fragmentation in mass spectrometers for protein identification, surveying of the different types of instrument configurations and their combinations for protein identification. The first mass spectrometer was developed in 1899, but it took almost a century for the instrument to become a routine analytical method in proteomic research when fast atom bombardment ionization was developed, followed shortly by soft desorption/ionization methods, such as MALDI and electrospray ionization, to volatize biomolecules with masses of tens of kiloDaltons into the gas phase under vacuum pressure without destroying them. Thereafter, other soft ionization techniques that offered ambient conditions were also introduced, such as atmospheric pressure MALDI, direct analysis in real time, atmospheric-pressure solid analysis probe and hybrid ionization, sources of MALDI and electrospray ionization (e.g., two-step fused droplet electrospray ionization, laser desorption atmospheric-pressure chemical ionization, electrosonic spray ionization, desorption electrospray ionization, and electrospray-assisted laser desorption/ionization). The five basic types of mass analyzers currently used in proteomic research are the quadrupole, ion trap, orbitrap, Fourier transform ion cyclotron resonance and TOF instruments, which differ in how they determine the mass-to-charge ratios of the peptides. They have very different design and performance characteristics. These analyzers can be stand alone or, in some cases, put together in tandem or in conjunction with ion mobility mass spectrometry to take advantage of the strengths of each. Several singly or multiply charged fragment ion types, such as b, y, a, c, z, v, y and immonium ions are produced in the gas phase of the spectrometer. In the bottom-up sequencing approach for protein identification in a shotgun proteomic experiment, proteolytic digestion of proteins is accomplished by cleavage of the different bonds along the peptide backbone and/or side chain through a charge-directed transfer to the vicinity of the cleavage side. These various mass spectrometers and the types of ions produced have become important analytical tools for studying and analyzing proteins, peptides and amino acids.
Ahmed
Farid E
FE
General, Environmental and Molecular Toxicology (GEM Tox) Consultants & Labs, Inc., 2607 Calvin Way, Greenville, NC 27834, USA. gemtoxconsultants@yahoo.com
D000818:Animals; D002853:Chromatography, Liquid; D019075:Electrophoresis, Capillary; D006801:Humans; D013058:Mass Spectrometry / Q000295:instrumentation* / Q000379:methods*; D009862:Online Systems; D020543:Proteome / Q000032:analysis* / Q000737:chemistry / Q000378:metabolism*
D020543:Proteome