Validation of the Oregon Scientific BPU 330 for self-monitoring of blood pressure according to the International Protocol.
OBJECTIVE
Extensive marketing of devices for self-measurement of blood pressure has created a need for purchasers to be able to satisfy themselves that such devices have been evaluated according to agreed criteria. The Oregon Scientific BPU 330 blood pressure monitor is an electronic device for upper arm measurement. This study assessed the accuracy of the Oregon Scientific BPU 330 blood pressure monitor according to the International Protocol by the Working Group on Blood Pressure Monitoring of the European Society of Hypertension for validation of blood pressure measuring devices.


METHODS
52 participants over 30 years of age were studied in the validation. Nine blood pressure measurements were taken alternately with a mercury sphygmomanometer by two observers, and by the supervisor, using the BPU 330 device. A total of 33 participants were selected for the analysis. The validation was divided into two phases. Phase 1 included 15 participants. If the device passed phase 1, 18 more participants were included. The 99 pairs of measurements were compared according to the International Protocol. The device was given a pass/fail recommendation based on its accuracy compared with the mercury standard (within 5, 10, and 15 mmHg), as well as the number met in the ranges specified by the International Protocol.


RESULTS
The mean and standard deviation of the difference between the mean of the observers and the BPU 330 device were 1.7 +/- 4.7 mmHg and 2.8 +/- 3.9 mmHg for systolic blood pressure (SBP) and diastolic blood pressure (DBP), respectively. In phase 1, the device passed with a total of 33, 43, and 44 SBP readings; 38, 44, and 45 DBP readings were within 5, 10, and 15 mmHg, respectively. In phase 2.1, 81, 95, and 96 for SBP, and 83, 95, and 98 for DBP readings fell within the zones of 5, 10, and 15 mmHg, respectively. In phase 2.2, the last phase, 28 participants fell within the zone of two of the three comparisons, lying within 5 mmHg for SBP and 29 participants for DBP. No participants fell within the zone of all three of their comparisons over 5 mmHg apart for both SBP and DBP.


CONCLUSIONS
The BPU 330 can be recommended for self-monitoring of blood pressure in the adult population, according to the International Protocol.
Li
Li
L
Biomedical Engineering Lab, Faculty of Information Engineering, ShenZhen University, ShenZhen, China.
Zhang
XinYu
X
Yan
ChunHong
C
Liang
QingXiang
Q
D000328:Adult; D000368:Aged; D001132:Arm / Q000098:blood supply*; D001794:Blood Pressure; D018660:Blood Pressure Monitoring, Ambulatory / Q000295:instrumentation* / Q000592:standards; D015924:Blood Pressure Monitors / Q000592:standards; D004867:Equipment Design; D005260:Female; D006801:Humans; D008297:Male; D008875:Middle Aged; D017410:Practice Guidelines as Topic; D011237:Predictive Value of Tests; D015203:Reproducibility of Results