title: Correlation between human immunodeficiency virus type 1 (HIV-1) RNA measurements obtained with dried blood spots and those obtained with plasma by use of Nuclisens EasyQ HIV-1 and Abbott RealTime HIV load tests.
abstract: The plasma human immunodeficiency virus (HIV) RNA load is used in the clinical routine for the monitoring of HIV infection and the patient's response to antiretroviral therapy. Other body fluids or dried blood spots (DBS) can be used, however, to assess the level of viremia. The use of DBS may be especially helpful for the monitoring of HIV-infected patients in resource-poor settings, where access to adequate laboratory facilities is often difficult. However, the correlation between the HIV RNA levels in plasma and those in DBSs has not been well established. Paired plasma and DBS samples obtained from HIV type 1 (HIV-1)-infected patients were tested for HIV RNA copy numbers by using two different commercial assays, the Nuclisens EasyQ HIV-1 (version 1.1) test (the Nuclisens test; Biomerieux) and the m2000rt RealTime HIV test (the m2000rt test; Abbott). Nucleic acid extraction was performed manually by using either the Nuclisens isolation kit (which uses the Boom methodology) or the m2000rt sample preparation kit (an iron particle-based method). A total of 103 paired plasma and DBS samples were tested. Viral load results were obtained for 97 (94.2%) samples with the Nuclisens isolation kit and 81 (78.6%) samples with the m2000rt kit. The overall correlation between the RNA loads in plasma and DBS was good, although better results were obtained by the Nuclisens test (R(2) = 0.87, P < 0.001) than by the m2000rt test (R(2) = 0.70, P < 0.001). While the specificities were excellent and similar for both the Nuclisens and the m2000rt tests (97.1% and 100%, respectively), the sensitivity was greater by the Nuclisens test than by the m2000rt test (75.8% and 56.6%, respectively). Overall, the viral loads in DBS tended to be lower than those in plasma, with mean differences of 0.3 log unit (standard deviation, 0.5 log unit) and 0.76 log unit (standard deviation, 0.8 log unit) for the Nuclisens and the m2000rt tests, respectively. The levels of agreement between the measurements in plasma and DBS were assessed by using the Bland-Altman plot for each assay. The Nuclisens test gave results within its defined limits (-0.65 to 1.26) for 95.9% of the samples, while the m2000rt test gave results within its limits (-0.83 to 2.33) for 100% of the samples. In summary, the HIV-1 load can accurately be quantified by testing DBS by either the Nuclisens or the m2000rt test, although the Nuclisens test may outperform the m2000rt test when nucleic acids are extracted manually.
authors.affiliation: Department of Infectious Diseases, Hospital Carlos III, Calle Sinesio Delgado 10, Madrid 28029, Spain. cmendoza@teleline.es.
authors.forename: Carolina
authors.initials: C
authors.lastname: Garrido
authors.forename: Natalia
authors.initials: N
authors.lastname: Zahonero
authors.forename: Angélica
authors.initials: A
authors.lastname: Corral
authors.forename: Miguel
authors.initials: M
authors.lastname: Arredondo
authors.forename: Vincent
authors.initials: V
authors.lastname: Soriano
authors.forename: Carmen
authors.initials: C
authors.lastname: de Mendoza
mesh_terms: D001769:Blood / Q000821:virology*; D003890:Desiccation; D015497:HIV-1 / Q000302:isolation & purification*; D006801:Humans; D025202:Molecular Diagnostic Techniques; D010949:Plasma / Q000821:virology*; D012367:RNA, Viral / Q000235:genetics; D011933:Reagent Kits, Diagnostic; D012680:Sensitivity and Specificity; D013048:Specimen Handling / Q000379:methods*; D019562:Viral Load
chemical_list: D012367:RNA, Viral; D011933:Reagent Kits, Diagnostic