Use of the labscale tangential flow diafiltrator to remove tetrameric hemoglobin from polyhemoglobin, purify hemolysate, and concentrate polyhemoglobin.
The Labscale tangential flow diafiltration system in conjunction with polyethersulfone Biomax filters (both purchased from Millipore Inc.) were used in the following studies: (1) A crucial step in the preparation of polyhemoglobin (polyHb) is the removal of tetrameric hemoglobin (Hb), which can cause adverse side-effects. The efficiency for this depends on the integrity of the 100kDa filters. Those with lower integrity were not effective whereas those with a higher integrity of 0.075ml/min were more effective. A filter with an integrity of 0.075ml/min can reduce the initial 11.2% tetrameric Hb concentration to 1.3% using a flow rate of 0.2ml/min. Higher flow rate of 2.2ml/min was not as effective. (2) Filtration with a 100kDa filter was quick and efficient in separating stroma-free hemoglobin solutions from cellular debris from hemolysate. (3) Both 5kDa and 100kDa filters are efficient in concentrating hemoglobin solutions.
Stefanescu
Ada
A
Artificial Cells & Organs Research Centre, Department of Physiology, McGill University, Montreal, Quebec, Canada.
Chang
Thomas Ming Swi
TM
D001802:Blood Substitutes / Q000737:chemistry / Q000302:isolation & purification / Q000378:metabolism; D005374:Filtration / Q000295:instrumentation / Q000379:methods*; D006454:Hemoglobins / Q000737:chemistry / Q000302:isolation & purification* / Q000378:metabolism; D006801:Humans; D011108:Polymers / Q000737:chemistry / Q000378:metabolism*; D055503:Protein Multimerization; D012106:Research / Q000295:instrumentation; D012107:Research Design; D013450:Sulfones / Q000737:chemistry / Q000378:metabolism*
D001802:Blood Substitutes; D006454:Hemoglobins; D011108:Polymers; D013450:Sulfones; C037158:polyhemoglobin; C022840:polyether sulfone