title: [Manufacture of high frequency direct current dental X-ray machine].
abstract: This paper introduces five dental X-ray machines which are manufactured with IGBT to realize high frequency, real-time sampling and PWM to ensure the closed-loop control for tube current and Anodes's high voltage. These five machines also use microcomputer and combined X-ray tube for precise control. The sets don't have high voltage outside. The error of tube voltage is less than 1% and exposure time is less than 3%. The photos of pulp cavity and surrounding tissue can be seen clearly. These sets surely meet the requirements of perspective use in clinical.
authors.affiliation: School of Mechanical Engineering, University of South China, Hengyang, Hunan 421001, China.
authors.forename: Jun-Jie
authors.initials: JJ
authors.lastname: Su
authors.forename: Hai-Hui
authors.initials: HH
authors.lastname: Chen
authors.forename: Yang-De
authors.initials: YD
authors.lastname: Zhang
mesh_terms: D004867:Equipment Design; D006801:Humans; D011856:Radiographic Image Enhancement; D011857:Radiographic Image Interpretation, Computer-Assisted / Q000379:methods*; D011861:Radiography, Dental / Q000295:instrumentation*