A diamond anvil cell for x-ray fluorescence measurements of trace elements in fluids at high pressure and high temperature.
We present a new diamond anvil cell (DAC), hereafter called the fluoX DAC, dedicated for x-ray fluorescence (XRF) analysis of trace elements in fluids under high pressure and high temperature to 10 GPa and 1273 K at least. This new setup has allowed measurement of Rb, Sr, Y, Zr, with concentrations of 50 ppm to 5.6 GPa and 1273 K. The characteristics of the fluoX DAC consist in an optimized shielding and collection geometry in order to reduce the background level in XRF spectrum. Consequently, minimum detection limits of 0.3 ppm were calculated for the abovementioned elements in this new setup. This new DAC setup coupled to the hard x-rays focusing beamline ID22 (ESRF, France) offers the possibility to analyze in situ at high pressure and high temperature, ppm level concentrations of heavy elements, rare earth elements, and first transition metals, which are of prime importance in geochemical processes. The fluoX DAC is also suitable to x-ray diffraction over the same high pressure-temperature range.
Petitgirard
Sylvain
S
Université de Lyon, F-69622 Lyon, France.
Daniel
Isabelle
I
Dabin
Yves
Y
Cardon
Hervé
H
Tucoulou
Rémi
R
Susini
Jean
J