Calibration beads containing luminescent lanthanide ion complexes.
The reliability of lanthanide luminescence measurements, by both flow cytometry and digital microscopy, would be enhanced by the availability of narrowband emitting, UV excited lanthanide calibration beads. 0.5-, 3-, and 5-microm beads containing a luminescent europium-complex are manufactured. The luminescence distribution of the 5-microm beads is measured with a time-delayed light-scatter-gated luminescence flow cytometer to have a 7.0% coefficient of variation (CV) The spacial distribution of the europium-complex in individual beads is determined to be homogeneous by confocal microscopy. Emission peaks are found at 592, 616 (width 9.9 nm), and 685 nm with a PARISS spectrophotometer. The kinetics of the luminescence bleaching caused by UV irradiation of the 0.5- and 5-microm beads measured under LED excitation with a fluorescence microscope indicate that bleaching does not interfere with their imaging. The luminescence lifetimes in water and air were 340 and 460 micros, respectively. Thus, these 5-microm beads can be used for spectral calibration of microscopes equipped with a spectrograph, as test particles for time-delayed luminescence flow cytometers, and possibly as labels for macromolecules and cells.
Leif
Robert C
RC
Newport Instruments, 5648 Toyon Road, San Diego, California 92115-1022, USA. rleif@rleif.com
Yang
Sean
S
Jin
Dayong
D
Piper
James
J
Vallarino
Lidia M
LM
Williams
John W
JW
Zucker
Robert M
RM
D002138:Calibration; D005434:Flow Cytometry / Q000592:standards*; D028581:Lanthanoid Series Elements / Q000032:analysis* / Q000737:chemistry*; D008163:Luminescent Measurements / Q000592:standards*; D008856:Microscopy, Fluorescence / Q000592:standards*; D008863:Microspheres; D013050:Spectrometry, Fluorescence / Q000592:standards*
D028581:Lanthanoid Series Elements