Mechanical characteristics of composite knitted stents.
We used metal wires and fibers to fabricate a composite knitted stent and then compare the mechanical characteristics of this stent with those of a pure metallic stent of the same construction in order to develop a stent that offers a comparable degree of expandability as metallic stents but can be used for highly curved lesions that cannot be treated using metallic stents. We fabricated two types of composite knitted stent (N-Z stents), using nitinol wire with a diameter of 0.12 mm and polypara-phenylene-benzobisoxazole (PBO) multifilament fiber (Zyron AS; Toyobo, Osaka, Japan). Stents were knitted into a cylindrical shape using the same textile pattern as a Strecker stent. Two loop lengths (L) of nitinol wire were used in the N-Z stents: L = 1.84 mm (N-Z stent L = 1.84) and L = 2.08 mm (N-Z stent L = 2.08). For the sake of comparison, we fabricated a metallic stent of nitinol using the same textile pattern (N-N stent L = 1.92). We applied a radial compression force diametrically to each stent and applied a bending force diametrically at the free end of a stent with one end fixed in order to evaluate the relationship between stent elasticity and load values. In addition, we macroscopically evaluated the generation of kinks when the stent was bent 180 degrees . The radial compressive force when the stent diameter was reduced by 53% was 6.44 N in the case of N-Z stent L = 1.84, 6.14 N in the case of N-Z stent L = 2.08, and 4.96 N in the case of N-N stent L = 1.92 mm. The composite stent had a radial compressive force higher than that of a metallic stent. The restoring force to longitudinal direction at a 90 degrees bending angle was 0.005 N for N-Z stent L = 1.84, 0.003 N for N-Z stent L = 2.08, and 0.034 N for N-N stent L = 1.92. The restoring force of the composite stent was significantly lower. Finally, the composite stent generated no definitive kinks at a bending angle of 180 degrees , regardless of loop length. However, the N-N stent clearly produced kinks, causing blockage of the inner cavity. In conclusion, the use of a metal and fiber composite in the construction of a knitted stent ensures an expansion performance comparable to that of metallic stents, while providing better kink resistance.
Tokuda
Takanori
T
Department of Radiology, Kansai Medical University Hirakata Hospital, Hirakata, Osaka, Japan. tkdtknr@gmail.com
Shomura
Yuzo
Y
Tanigawa
Noboru
N
Kariya
Shuji
S
Komemushi
Atsushi
A
Kojima
Hiroyuki
H
Sawada
Satoshi
S
D000497:Alloys / Q000737:chemistry*; D001672:Biocompatible Materials; D019245:Compressive Strength; D006801:Humans; D008422:Materials Testing; D011108:Polymers / Q000737:chemistry*; D011474:Prosthesis Design; D015607:Stents; D013718:Tensile Strength
D000497:Alloys; D001672:Biocompatible Materials; D011108:Polymers; C013616:nitinol