title: Revision characteristics of cement-augmented, cannulated-fenestrated pedicle screws in the osteoporotic vertebral body: a biomechanical in vitro investigation. Technical note.
abstract: In generalized osteoporosis, instrumentation with cement-augmented pedicle screws is an amplification of the therapeutic spectrum. Early clinical results are promising for both solid and cannulated screws; however, there are concerns regarding the revision characteristics of these screws, especially for the cannulated-fenestrated type with its continuous cement interconnection from the core of the screw to surrounding bone tissue. In a human cadaver model, bone mineral density (BMD) was assessed radiographically. Spinal levels T9-L4 were instrumented left unilaterally, transpedicularly by using cannulated-fenestrated pedicle screws with the dimensions 6.5 x 45 mm. Polymethylmethacrylate cement (1.5 ml) was injected through the screws into each vertebra. After polymerization of the cement, the extraction torque was recorded. For both implantation and explantation of the screws, a fluoroscope was used to guarantee correct screw and cement positioning and to observe possible co-movements-that is, any movement of the cement mass within the vertebral body upon removal of the screw. For comparison, the extraction torque of same-dimension pedicle screws was recorded in a nonosteoporotic, non-cement-augmented instrumentation. The BMD was 0.60 g/cm2, a level that corresponds to a severe grade of osteoporosis. For removal of the screws, the median and mean extraction torques were 34 and 49 +/- 44 Ncm, respectively. No co-movements of the cement mass occurred within the vertebral body. In the nonosteoporotic control, BMD was 1.38 g/cm2. The median and mean extraction torques were 123 and 124 +/- 12 Ncm, respectively. Thus, the revision characteristics of cement-augmented, cannulated-fenestrated pedicle screws are not problematic, even in cases of severe osteoporosis. The winglike cement interconnection between the screw core and surrounding bone tissue is fragile enough to break off in the event of an extraction torque and to release the screw. There is no proof to support the theoretical fear that while trying to remove a screw, the composite of screw and cement would not break but instead would rotate as a whole in the osteoporotic vertebral body.
authors: Clinic for Trauma, Reconstructive, and Plastic Surgery, Spine Center, University Hospital Leipzig, Germany. thomas.blattert@medizin.uni-leipzig.de
authors: Thomas R
authors: TR
authors: Blattert
authors: Stefan
authors: S
authors: Glasmacher
authors: Hans-Joachim
authors: HJ
authors: Riesner
authors: Christoph
authors: C
authors: Josten
mesh_terms: D001696:Biomechanical Phenomena; D001843:Bone Cements; D015519:Bone Density; D001863:Bone Screws; D002102:Cadaver; D005260:Female; D006801:Humans; D066298:In Vitro Techniques; D010024:Osteoporosis / Q000601:surgery*; D019904:Polymethyl Methacrylate; D011474:Prosthesis Design; D011859:Radiography; D012086:Reoperation; D013131:Spine / Q000000981:diagnostic imaging / Q000601:surgery*; D018709:Statistics, Nonparametric; D019415:Torque
chemical_list: D001843:Bone Cements; D019904:Polymethyl Methacrylate