The principle of low frictional torque in the Charnley total hip replacement.
The design of the Charnley total hip replacement follows the principle of low frictional torque. It is based on the largest possible difference between the radius of the femoral head and that of the outer aspect of the acetabular component. The aim is to protect the bone-cement interface by movement taking place at the smaller radius, the articulation. This is achieved in clinical practice by a 22.225 mm diameter head articulating with a 40 mm or 43 mm diameter acetabular component of ultra-high molecular weight polyethylene. We compared the incidence of aseptic loosening of acetabular components with an outer diameter of 40 mm and 43 mm at comparable depths of penetration with a mean follow-up of 17 years (1 to 40). In cases with no measurable wear none of the acetabular components were loose. With increasing acetabular penetration there was an increased incidence of aseptic loosening which reflected the difference in the external radii, with 1.5% at 1 mm, 8.8% at 2 mm, 9.7% at 3 mm and 9.6% at 4 mm of penetration in favour of the larger 43 mm acetabular component. Our findings support the Charnley principle of low frictional torque. The level of the benefit is in keeping with the predicted values.
Wroblewski
B M
BM
The John Charnley Research Institute, Wrightington Hospital, Hall Lane, Appley Bridge, Wigan, WN6 0TS, UK.
Siney
P D
PD
Fleming
P A
PA
D000077:Acetabulum / Q000503:physiopathology / Q000601:surgery*; D000293:Adolescent; D000328:Adult; D019644:Arthroplasty, Replacement, Hip / Q000009:adverse effects* / Q000379:methods; D001696:Biomechanical Phenomena / Q000502:physiology; D005260:Female; D017276:Friction / Q000502:physiology; D006621:Hip Joint / Q000503:physiopathology / Q000601:surgery*; D006622:Hip Prosthesis / Q000009:adverse effects*; D006801:Humans; D008297:Male; D008875:Middle Aged; D011474:Prosthesis Design; D011475:Prosthesis Failure; D019415:Torque; D055815:Young Adult