title: A practical approach for electron monitor unit calculation.
abstract: Electron monitor unit (MU) calculation requires measured beam data such as the relative output factor (ROF) of a cone, insert correction factor (ICF) and effective source-to-surface distance (ESD). Measuring the beam data to cover all possible clinical cases is not practical for a busy clinic because it takes tremendous time and labor. In this study, we propose a practical approach to reduce the number of data measurements without affecting accuracy. It is based on two findings of dosimetric properties of electron beams. One is that the output ratio of two inserts is independent of the cone used, and the other is that ESD is a function of field size but independent of cone and jaw opening. For the measurements to prove the findings, a parallel plate ion chamber (Markus, PTW 23343) with an electrometer (Cardinal Health 35040) was used. We measured the outputs to determine ROF, ICF and ESD of different energies (5-21 MeV). Measurements were made in a Plastic Water phantom or in water. Three linear accelerators were used: Siemens MD2 (S/N 2689), Siemens Primus (S/N 3305) and Varian Clinic 21-EX (S/N 1495). With these findings, the number of data set to be measured can be reduced to less than 20% of the data points.
authors.affiliation: Department of Radiation Medicine, Loma Linda University Medical Center, Loma Linda, CA, USA. dchoi@llurm.org
authors.forename: David
authors.initials: D
authors.lastname: Choi
authors.forename: Baldev
authors.initials: B
authors.lastname: Patyal
authors.forename: Jongmin
authors.initials: J
authors.lastname: Cho
authors.forename: Ing Y
authors.initials: IY
authors.lastname: Cheng
authors.forename: Prashanth
authors.initials: P
authors.lastname: Nookala
mesh_terms: D004583:Electrons; D011874:Radiometry / Q000379:methods*; D014867:Water
chemical_list: D014867:Water