A method to generate equivalent energy spectra and filtration models based on measurement for multidetector CT Monte Carlo dosimetry simulations.
The purpose of this study was to present a method for generating x-ray source models for performing Monte Carlo (MC) radiation dosimetry simulations of multidetector row CT (MDCT) scanners. These so-called "equivalent" source models consist of an energy spectrum and filtration description that are generated based wholly on the measured values and can be used in place of proprietary manufacturer's data for scanner-specific MDCT MC simulations. Required measurements include the half value layers (HVL1 and HVL2) and the bowtie profile (exposure values across the fan beam) for the MDCT scanner of interest. Using these measured values, a method was described (a) to numerically construct a spectrum with the calculated HVLs approximately equal to those measured (equivalent spectrum) and then (b) to determine a filtration scheme (equivalent filter) that attenuates the equivalent spectrum in a similar fashion as the actual filtration attenuates the actual x-ray beam, as measured by the bowtie profile measurements. Using this method, two types of equivalent source models were generated: One using a spectrum based on both HVL1 and HVL2 measurements and its corresponding filtration scheme and the second consisting of a spectrum based only on the measured HVL1 and its corresponding filtration scheme. Finally, a third type of source model was built based on the spectrum and filtration data provided by the scanner's manufacturer. MC simulations using each of these three source model types were evaluated by comparing the accuracy of multiple CT dose index (CTDI) simulations to measured CTDI values for 64-slice scanners from the four major MDCT manufacturers. Comprehensive evaluations were carried out for each scanner using each kVp and bowtie filter combination available. CTDI experiments were performed for both head (16 cm in diameter) and body (32 cm in diameter) CTDI phantoms using both central and peripheral measurement positions. Both equivalent source model types result in simulations with an average root mean square (RMS) error between the measured and simulated values of approximately 5% across all scanner and bowtie filter combinations, all kVps, both phantom sizes, and both measurement positions, while data provided from the manufacturers gave an average RMS error of approximately 12% pooled across all conditions. While there was no statistically significant difference between the two types of equivalent source models, both of these model types were shown to be statistically significantly different from the source model based on manufacturer's data. These results demonstrate that an equivalent source model based only on measured values can be used in place of manufacturer's data for Monte Carlo simulations for MDCT dosimetry.
Turner
Adam C
AC
Department of Biomedical Physics and Department of Radiology, David Geffen School of Medicine, University of California, Los Angeles, Los Angeles, California 90024, USA. aturner@mednet.ucla.edu
Zhang
Di
D
Kim
Hyun J
HJ
DeMarco
John J
JJ
Cagnon
Chris H
CH
Angel
Erin
E
Cody
Dianna D
DD
Stevens
Donna M
DM
Primak
Andrew N
AN
McCollough
Cynthia H
CH
McNitt-Gray
Michael F
MF
D000465:Algorithms; D001822:Body Burden; D003198:Computer Simulation; D005374:Filtration / Q000379:methods*; D006801:Humans; D008954:Models, Biological; D009010:Monte Carlo Method; D011874:Radiometry / Q000379:methods*; D012062:Relative Biological Effectiveness; D012542:Scattering, Radiation; D014057:Tomography, X-Ray Computed / Q000379:methods*