title: Characterization of the sunset semi-continuous carbon aerosol analyzer.
abstract: The field-deployable Sunset Semi-Continuous Organic Carbon/Elemental Carbon (Sunset OCEC) aerosol analyzer utilizes the modified National Institute for Occupational Safety and Health thermal-optical method to determine total carbon (TC), organic carbon (OC), and elemental carbon (EC) at near real-time. Two sets of OC and EC are available: thermal OC and EC, and optical OC and EC. The former is obtained by the thermal-optical approach, and the latter is obtained by directly determining EC optically and deriving optical OC from TC. However, the performance of the Sunset OCEC is not yet fully characterized. Two collocated Sunset OCEC analyzers, Unit A and Unit B, were used to determine the pooled relative standard deviation (RSD) and limit of detection (LOD) between September 18 and November 6, 2007 in Richland, WA. The LOD of Unit A was approximately 0.2 microgC/m3 (0.1 microgC/cm2) for TC, optical OC, and thermal OC, and 0.01 microgC/m3 (0.01 microgC/cm2) for optical EC. Similarly, Unit B had an LOD of approximately 0.3 microgC/m3 (0.2 microgC/cm2) for TC, optical OC, and thermal OC, and 0.02 microgC/m3 (0.01 microgC/cm2) for optical EC. The LOD for thermal EC is estimated to be 0.2 microgC/m3 (0.1 microgC/cm2) for both units. The pooled RSDs were 4.9% for TC (carbon mass loadings 0.6-6.0 microgC/cm2), 5.6% for optical OC (carbon mass loadings 0.6-5.4 microgC/cm2), 5.3% for thermal OC (carbon mass loadings 0.6-5.3 microgC/ cm2), and 9.6% for optical EC (carbon mass loadings 0-1.4 microgC/cm2), which indicates good precision between the instruments. The RSD for thermal EC is higher at 24.3% (carbon mass loadings 0-1.2 microgC/cm2). Low EC mass loadings in Richland contributed to the poor RSD of EC. The authors found that excessive noise from the nondispersive infrared (NDIR) laser in the Sunset OCEC analyzer could result in a worsened determination of OC and EC. It is recommended that a "quieter" NDIR laser and detector be used in the Sunset OCEC analyzer to improve quantification. Future work should re-evaluate the precision of the EC parameters in an environment favorable for EC collection. Investigation among quantification differences using various thermal-optical protocols to determine OC and EC is also in need.
authors: Department of Earth and Atmospheric Sciences, Purdue University, West Lafayette, IN, USA.
authors: Jace J
authors: JJ
authors: Bauer
authors: Xiao-Ying
authors: XY
authors: Yu
authors: Robert
authors: R
authors: Cary
authors: Nels
authors: N
authors: Laulainen
authors: Carl
authors: C
authors: Berkowitz
mesh_terms: D000336:Aerosols / Q000032:analysis*; D000393:Air Pollutants / Q000032:analysis* / Q000737:chemistry; D002244:Carbon / Q000032:analysis*; D004784:Environmental Monitoring / Q000295:instrumentation*; D052638:Particulate Matter / Q000032:analysis* / Q000737:chemistry; D015203:Reproducibility of Results
chemical_list: D000336:Aerosols; D000393:Air Pollutants; D052638:Particulate Matter; D002244:Carbon