Microfluidic blood plasma separation via bulk electrohydrodynamic flows.
An effective mechanism for rapid and efficient microfluidic particle trapping and concentration is proposed without requiring any mechanically moving parts. When a voltage beyond the threshold atmospheric ionization value is applied on a sharp electrode tip mounted at an angle above a microfluidic liquid chamber, the bulk electrohydrodynamic air thrust that is generated results in interfacial shear and, hence, primary azimuthal liquid surface recirculation. This discharge driven vortex mechanism, in turn, causes a secondary bulk meridional liquid recirculation, which produces an inward radial force near the bottom of the chamber. Particles suspended in the liquid are then rapidly convected by the bulk recirculation toward the bottom, where the inward radial force causes them to spiral in a helical swirl-like fashion toward a stagnation point. In particular, we show that these flows, similar to Batchelor flows occurring in a cylindrical liquid column between a stationary and rotating disk, can be used for the separation of red blood cells from blood plasma in a miniaturized device.
Arifin
Dian R
DR
MicroNanophysics Research Laboratory, Department of Mechanical Engineering,Monash University, Clayton, VIC 3800, Australia.
Yeo
Leslie Y
LY
Friend
James R
JR
Electrophoresis. 2002 Jul;23(13):1973-83
12210248
Electrophoresis. 2003 Nov;24(21):3703-17
14613196
Lab Chip. 2006 Jun;6(6):776-81
16738730
Clin Chim Acta. 1960 Sep;5:719-26
13788465
Biotechnol Prog. 2003 Nov-Dec;19(6):1798-811
14656159
Microvasc Res. 1973 Jan;5(1):34-48
4684755
Conf Proc IEEE Eng Med Biol Soc. 2004;2004:2627-30
17270814
Genome Res. 2001 Mar;11(3):405-12
11230164
ASAIO J. 2005 Sep-Oct;51(5):585-90
16322722
Lab Chip. 2006 Jul;6(7):871-80
16804591
Clin Chem. 1994 Jan;40(1):43-7
8287542
Annu Rev Biomed Eng. 2005;7:77-103
16004567
J Biomed Mater Res. 1999 Mar 5;44(3):289-97
10397931
Clin Chim Acta. 1961 Jul;6:538-44
14453500