Ultrasound phase contrast thermal imaging with reflex transmission imaging methods in tissue phantoms.
Thermal imaging measurements using ultrasound phase contrast have been performed in tissue phantoms heated with a focused ultrasound source. Back projection and reflex transmission imaging principles were used to detect sound speed-induced changes in the phase caused by an increase in the temperature. The temperature was determined from an empirical relationship for the temperature dependence on sound speed. The phase contrast was determined from changes in the sound field measured with a hydrophone scan conducted before and during applied heating. The lengthy scanning routine used to mimic a large two-dimensional array required a steady-state temperature distribution within the phantom. The temperature distribution in the phantom was validated with magnetic resonance (MR) thermal imaging measurements. The peak temperature was found to agree within 1 degrees C with MR, and good agreement was found between the temperature profiles. The spatial resolution was 0.3x0.3x0.3mm, comparing favorably with the 0.625x0.625x1.5-mm MR spatial resolution.
Farny
Caleb H
CH
Department of Radiology, Harvard Medical School, Brigham and Women's Hospital, Boston, MA 02115, USA. cfarny@bwh.harvard.edu
Clement
Gregory T
GT
D006801:Humans; D007090:Image Interpretation, Computer-Assisted; D019047:Phantoms, Imaging; D012031:Refractometry / Q000379:methods*; D015203:Reproducibility of Results; D012680:Sensitivity and Specificity; D013696:Temperature; D013817:Thermography / Q000379:methods*; D014463:Ultrasonography / Q000295:instrumentation / Q000379:methods*
J Acoust Soc Am. 1997 Jul;102(1):628-34
9228822
Phys Med Biol. 2003 Jan 21;48(2):223-41
12587906
Ultrasound Med Biol. 2002 Oct;28(10):1319-33
12467859
IEEE Trans Ultrason Ferroelectr Freq Control. 1998;45(4):1088-99
18244264
Ultrason Imaging. 1985 Jul;7(3):201-14
3913096
Magn Reson Med. 1995 Dec;34(6):814-23
8598808
Int J Hyperthermia. 2005 Sep;21(6):533-46
16147438
Ultrasound Med Biol. 1979;5(2):149-57
505616
Phys Med Biol. 2005 Apr 7;50(7):1585-600
15798345
Magn Reson Med. 2003 Aug;50(2):322-30
12876709
J Magn Reson Imaging. 2008 Feb;27(2):376-90
18219673
Ultrasound Med Biol. 2003 Feb;29(2):293-9
12659917
Ultrasound Med Biol. 2002 Mar;28(3):321-9
11978412
IEEE Trans Ultrason Ferroelectr Freq Control. 2001 Nov;48(6):1686-94
11800132
Ultrasound Med Biol. 1985 Mar-Apr;11(2):293-8
4035805
IEEE Trans Ultrason Ferroelectr Freq Control. 2004 May;51(5):606-15
15217237
Int J Hyperthermia. 2003 Nov-Dec;19(6):617-41
14756452
Ultrasound Med Biol. 2008 Sep;34(9):1449-64
18450361
Int J Hyperthermia. 1996 Sep-Oct;12(5):573-91; discussion 593-4
8886886
J Acoust Soc Am. 1984 Jun;75(6):1732-40
6747093
J Acoust Soc Am. 2004 Sep;116(3):1819-25
15478449
IEEE Trans Ultrason Ferroelectr Freq Control. 2007 Jan;54(1):23-31
17225797
Ultrasound Med Biol. 2003 Jan;29(1):127-35
12604124
Ultrasound Med Biol. 2004 Mar;30(3):345-56
15063516
Magn Reson Med. 2003 Jan;49(1):89-98
12509823
J Invest Dermatol. 2007 Jan;127(1):189-95
17068484
Ultrasound Med Biol. 2008 Mar;34(3):414-24
18187251
Ultrasound Med Biol. 1991;17(3):283-9
1887513
Ultrasound Med Biol. 1997;23(2):261-73
9140183
J Magn Reson Imaging. 2000 Oct;12(4):525-33
11042633
Phys Med Biol. 2007 Aug 21;52(16):4827-43
17671338
Ultrasound Med Biol. 2001 May;27(5):673-82
11397532
J Acoust Soc Am. 1972 Apr;51(4):1333-51
5032950
Phys Med Biol. 2004 Feb 21;49(4):533-46
15005163
IEEE Trans Ultrason Ferroelectr Freq Control. 1993;40(2):84-102
18263161
J Acoust Soc Am. 2008 Mar;123(3):1773-83
18345865
J Acoust Soc Am. 1996 Oct;100(4 Pt 1):2522-30
8865655
Ultrason Imaging. 2004 Jul;26(3):185-200
15754799
MAGMA. 2001 Aug;13(1):47-59
11410396