Urgent exchange of a HeartMate II left ventricular assist device after percutaneous lead fracture.
In four of our patients implanted with the HeartMate II left ventricular assist device (LVAD), fraying and fracture of the percutaneous lead's internal wires led to pump failure. The patients included two adolescent males (age, 15 and 17 years), one adult male (age, 78 years), and one adult female (age, 49 years). The damage was attributed to patient-related twisting or kinking of the lead. The patients had been supported by the LVAD for 6-18 months before the lead was damaged. In each case, audible and visual alarms were activated, and the pump temporarily continued to function well. The patients were rushed to the hospital, and the pumps were exchanged. All four patients recovered uneventfully and were discharged from the hospital 10-14 days later. They continued to do well on LVAD support for up to 639 days after the initial implant procedure. These events show the importance of educating LVAD patients to avoid excessive twisting or kinking of the device's driveline.
Jafar
Mehdi
M
Center for Cardiac Support, Texas Heart Institute at St. Luke's Episcopal Hospital, Houston, Texas 77225-0345, USA.
Gregoric
Igor D
ID
Radovancevic
Rajko
R
Cohn
William E
WE
McGuire
Nichole
N
Frazier
O H
OH
D000293:Adolescent; D000368:Aged; D004868:Equipment Failure; D005260:Female; D006353:Heart-Assist Devices / Q000009:adverse effects*; D006801:Humans; D008297:Male; D008875:Middle Aged