Monitoring with head-mounted displays: performance and safety in a full-scale simulator and part-task trainer.
BACKGROUND
Head-mounted displays (HMDs) can help anesthesiologists with intraoperative monitoring by keeping patients' vital signs within view at all times, even while the anesthesiologist is busy performing procedures or unable to see the monitor. The anesthesia literature suggests that there are advantages of HMD use, but research into head-up displays in the cockpit suggests that HMDs may exacerbate inattentional blindness (a tendency for users to miss unexpected but salient events in the field of view) and may introduce perceptual issues relating to focal depth. We investigated these issues in two simulator-based experiments.


METHODS
Experiment 1 investigated whether wearing a HMD would affect how quickly anesthesiologists detect events, and whether the focus setting of the HMD (near or far) makes any difference. Twelve anesthesiologists provided anesthesia in three naturalistic scenarios within a simulated operating theater environment. There were 24 different events that occurred either on the patient monitor or in the operating room. Experiment 2 investigated whether anesthesiologists physically constrained by performing a procedure would detect patient-related events faster with a HMD than without. Twelve anesthesiologists performed a complex simulated clinical task on a part-task endoscopic dexterity trainer while monitoring the simulated patient's vital signs. All participants experienced four different events within each of two scenarios.


RESULTS
Experiment 1 showed that neither wearing the HMD nor adjusting the focus setting reduced participants' ability to detect events (the number of events detected and time to detect events). In general, participants spent more time looking toward the patient and less time toward the anesthesia machine when they wore the HMD than when they used standard monitoring alone. Participants reported that they preferred the near focus setting. Experiment 2 showed that participants detected two of four events faster with the HMD, but one event more slowly with the HMD. Participants turned to look toward the anesthesia machine significantly less often when using the HMD. When using the HMD, participants reported that they were less busy, monitoring was easier, and they believed they were faster at detecting abnormal changes.


CONCLUSIONS
The HMD helped anesthesiologists detect events when physically constrained, but not when physically unconstrained. Although there was no conclusive evidence of worsened inattentional blindness, found in aviation, the perceptual properties of the HMD display appear to influence whether events are detected. Anesthesiologists wearing HMDs should self-adjust the focus to minimize eyestrain and should be aware that some changes may not attract their attention. Future areas of research include developing principles for the design of HMDs, evaluating other types of HMDs, and evaluating the HMD in clinical contexts.
Liu
David
D
School of Information Technology and Electrical Engineering, The University of Queensland, Brisbane, Australia. naskies@acm.org
Jenkins
Simon A
SA
Sanderson
Penelope M
PM
Watson
Marcus O
MO
Leane
Terrence
T
Kruys
Amanda
A
Russell
W John
WJ
D000776:Anesthesiology / Q000295:instrumentation*; D001288:Attention; D001307:Auditory Perception; D001364:Awareness; D002983:Clinical Competence; D003463:Cues; D003626:Data Display; D004724:Endoscopy; D004867:Equipment Design; D006257:Head; D006801:Humans; D008348:Manikins; D016343:Monitoring, Intraoperative / Q000295:instrumentation*; D009048:Motor Skills; D009873:Operating Rooms; D011930:Reaction Time; D011795:Surveys and Questionnaires; D013647:Task Performance and Analysis; D014792:Visual Acuity; D014796:Visual Perception