Distribution of periacetabular osteolytic lesions varies according to component design.
Using computed tomography, the volume, location, and number of osteolytic lesions were determined adjacent to 38 Harris-Galante 1 (HG-1) acetabular components fixed with screws and 19 porous-coated anatomic (PCA) acetabular components press-fitted without screws. The median implantation times were 16 and 15 years, respectively. The mean total lesion volumes were similar: 11.1 cm(3) (range, 0.7-49 cm(3)) and 9.8 cm(3) (range, 0.4-52 cm(3)), respectively, for hips with HG-1 and PCA components (P = .32). There was a significant difference in the proportion of rim-related, screw or screw hole-related, and combined lesions between the 2 component designs (P < .0001). HG-1 components had more screw and screw hole-related lesions, and PCA components had more rim-related lesions. Although there are concerns regarding screw and screw hole-associated osteolysis, these findings suggest that peripheral fixation may be well maintained in the long term with the use of multiple-hole acetabular components with screw fixation.
Stamenkov
Roumen B
RB
Department of Orthopaedics and Trauma, Royal Adelaide Hospital, Adelaide, South Australia.
Howie
Donald W
DW
Neale
Susan D
SD
McGee
Margaret A
MA
Taylor
David J
DJ
Findlay
David M
DM
D000077:Acetabulum / Q000000981:diagnostic imaging; D000328:Adult; D000368:Aged; D019644:Arthroplasty, Replacement, Hip; D005260:Female; D006621:Hip Joint / Q000000981:diagnostic imaging*; D006622:Hip Prosthesis / Q000009:adverse effects*; D006801:Humans; D008297:Male; D008875:Middle Aged; D010014:Osteolysis / Q000000981:diagnostic imaging* / Q000209:etiology; D011474:Prosthesis Design; D011475:Prosthesis Failure; D012086:Reoperation; D014057:Tomography, X-Ray Computed