Split gradient coils for simultaneous PET-MRI.
Combining positron emission tomography (PET) and MRI necessarily involves an engineering tradeoff as the equipment needed for the two modalities vies for the space closest to the region where the signals originate. In one recently described scanner configuration for simultaneous positron emission tomography-MRI, the positron emission tomography detection scintillating crystals reside in an 80-mm gap between the 2 halves of a 1-T split-magnet cryostat. A novel set of gradient and shim coils has been specially designed for this split MRI scanner to include an 110-mm gap from which wires are excluded so as not to interfere with positron detection. An inverse boundary element method was necessarily employed to design the three orthogonal, shielded gradient coils and shielded Z0 shim coil. The coils have been constructed and tested in the hybrid positron emission tomography-MRI system and successfully used in simultaneous positron emission tomography-MRI experiments.
Sir Peter Mansfield Magnetic Resonance Centre, School of Physics and Astronomy, University of Nottingham, University Park, Nottingham, Nottinghamshire, United Kingdom. michael@itee.uq.edu.au
Michael
M
Poole
Richard
R
Bowtell
Dan
D
Green
Simon
S
Pittard
Alun
A
Lucas
Rob
R
Hawkes
Adrian
A
Carpenter
D017076:Computer-Aided Design; D004867:Equipment Design; D019544:Equipment Failure Analysis; D007089:Image Enhancement / Q000295:instrumentation*; D008279:Magnetic Resonance Imaging / Q000295:instrumentation*; D008280:Magnetics / Q000295:instrumentation*; D019047:Phantoms, Imaging; D049268:Positron-Emission Tomography / Q000295:instrumentation*; D015203:Reproducibility of Results; D012680:Sensitivity and Specificity; D013382:Subtraction Technique / Q000295:instrumentation*; D014159:Transducers