[Evaluation of two automated methods for blood lactate levels].
We evaluated two automated methods for measuring blood levels of lactates with Integra 800 (Roche Diagnostics) and ABL 725 (Radiometer). Our evaluation had shown a within run imprecision of 1% (Intégra 800) and 2% (ABL 725) and a between-assay imprecision of 2% (Intégra 800) and 5% (ABL 725). The two methodologies appeared very well associated. The selective electrode remains expensive but it is very interesting because it saves a blood sample.
Désidéri-Vaillant
C
C
Fédération des laboratoires,, Service de réanimation, Hôpital d'instruction des armées Clermont Tonnerre, Brest.
Cardon
N
N
Giacardi
C
C
Valéro
E
E
Savin
P
P
Lebras
J
J
Corbé
H
H
D001774:Blood Chemical Analysis / Q000295:instrumentation*; D006801:Humans; D019344:Lactic Acid / Q000097:blood*
D019344:Lactic Acid