Sleep stage classification with low complexity and low bit rate.
Standard sleep stage classification is based on visual analysis of central (usually also frontal and occipital) EEG, two-channel EOG, and submental EMG signals. The process is complex, using multiple electrodes, and is usually based on relatively high (200-500 Hz) sampling rates. Also at least 12 bit analog to digital conversion is recommended (with 16 bit storage) resulting in total bit rate of at least 12.8 kbit/s. This is not a problem for in-house laboratory sleep studies, but in the case of online wireless self-applicable ambulatory sleep studies, lower complexity and lower bit rates are preferred. In this study we further developed earlier single channel facial EMG/EOG/EEG-based automatic sleep stage classification. An algorithm with a simple decision tree separated 30 s epochs into wakefulness, SREM, S1/S2 and SWS using 18-45 Hz beta power and 0.5-6 Hz amplitude. Improvements included low complexity recursive digital filtering. We also evaluated the effects of a reduced sampling rate, reduced number of quantization steps and reduced dynamic range on the sleep data of 132 training and 131 testing subjects. With the studied algorithm, it was possible to reduce the sampling rate to 50 Hz (having a low pass filter at 90 Hz), and the dynamic range to 244 microV, with an 8 bit resolution resulting in a bit rate of 0.4 kbit/s. Facial electrodes and a low bit rate enables the use of smaller devices for sleep stage classification in home environments.
Virkkala
Jussi
J
Sleep Laboratory, Finnish Institute of Occupational Health, Helsinki, Finland. jussi.virkkala@ttl.fi
Värri
Alpo
A
Hasan
Joel
J
Himanen
Sari-Leena
SL
Müller
Kiti
K
D000465:Algorithms; D003198:Computer Simulation; D004566:Electrodes; D004569:Electroencephalography / Q000379:methods*; D004576:Electromyography / Q000379:methods*; D004585:Electrooculography / Q000295:instrumentation / Q000379:methods*; D005583:Fourier Analysis; D006801:Humans; D010363:Pattern Recognition, Automated; D017286:Polysomnography / Q000379:methods; D012815:Signal Processing, Computer-Assisted; D012890:Sleep; D012894:Sleep Stages; D012984:Software; D014851:Wakefulness