Novel epicardial off-pump device for mitral regurgitation: acute evaluation.
OBJECTIVE
This study evaluates the ability of a novel epicardial annuloplasty device Mitral Touch (MAQUET Cardiovascular LLC, San Jose, CA, USA) to reduce functional mitral regurgitation (MR) in a rapid ventricular pacing-induced dilated cardiomyopathy model in dogs.


METHODS
A median sternotomy was performed in 13 dogs after MR induction by rapid ventricular pacing (230 beats/min for an average of 35.6 + or - 12.8 days). Two-dimensional epicardial echocardiographic and haemodynamic measurements were performed to evaluate the baseline MR grade, the septal-lateral (S-L) dimension of the mitral annulus, mitral valve (MV) geometry and left ventricular function. The Mitral Touch was implanted by sliding the anterior arm onto the floor of the transverse sinus and positioning the posterior arm just apical to the atrioventricular groove on the left ventricular posterolateral wall. The 24-mm-long device was implanted in eight dogs, the 27-mm-long device in four and the 30-mm standard length device in one. MR grade, S-L dimension and haemodynamics data acquisition were immediately rechecked after device implantation.


RESULTS
All implantations, which took only approximately 30s to deploy, were performed on beating hearts without cardiopulmonary bypass. In one early case, after extended manipulation with undersized devices, an atrial laceration was created and bleeding occurred. Design changes were made to eliminate this complication. The MR grade was significantly (p=0.003) reduced from 3.1 + or - 1.1 at baseline to 1.4 + or - 0.8 after device implantation. The S-L dimension at end of systole was also significantly (p=0.001) reduced from 2.7 + or - 0.4 cm at baseline to 2.3 + or - 0.3 cm after device implantation (% reduction: 15.1 + or - 10.6%). The mitral valve coaptation length was significantly (p=0.0001) increased from 0.36 + or - 0.11 cm to 0.50 + or - 0.08 cm, and the mitral valve tethering area was significantly (p=0.0003) decreased from 1.36 + or - 0.38 cm(2) to 0.81 + or - 0.29 cm(2) after Mitral Touch implantation.


CONCLUSIONS
This new epicardial device was effective in significantly reducing MR and S-L dimensions acutely on the beating heart without requiring the use of cardiopulmonary bypass. Further studies are necessary to confirm the long-term maintenance of MR and S-L reductions.
Takaseya
Tohru
T
Department of Biomedical Engineering, Lerner Research Institute, Cleveland Clinic, 9500 Euclid Avenue, Cleveland, OH 44195, USA.
Shiose
Akira
A
Saraiva
Roberto M
RM
Fumoto
Hideyuki
H
Arakawa
Yoko
Y
Juravic
Mark
M
Lombardi
Pierluca
P
Fukamachi
Kiyotaka
K
D000818:Animals; D002311:Cardiomyopathy, Dilated / Q000150:complications; D002315:Cardiopulmonary Bypass; D004195:Disease Models, Animal; D004285:Dogs; D006350:Heart Valve Prosthesis; D019918:Heart Valve Prosthesis Implantation / Q000295:instrumentation / Q000379:methods; D019060:Minimally Invasive Surgical Procedures / Q000295:instrumentation / Q000379:methods; D058386:Mitral Valve Annuloplasty / Q000295:instrumentation* / Q000379:methods; D008944:Mitral Valve Insufficiency / Q000000981:diagnostic imaging / Q000209:etiology / Q000601:surgery*; D011474:Prosthesis Design; D014463:Ultrasonography