Framework conditions and requirements to ensure the technical functional safety of reprocessed medical devices.
Testing and restoring technical-functional safety is an essential part of medical device reprocessing. Technical functional tests have to be carried out on the medical device in the course of the validation of reprocessing procedures. These ensure (in addition to the hygiene tests) that the reprocessing procedure is suitable for the medical device. Functional tests are, however, also a part of reprocessing procedures. As a stage in the reprocessing, they ensure for the individual medical device that no damage or other changes limit the performance. When determining which technical-functional tests are to be carried out, the current technological standard has to be taken into account in the form of product-specific and process-oriented norms. Product-specific norms primarily define safety-relevant requirements. The risk management method described in DIN EN ISO 14971 is the basis for recognising hazards; the likelihood of such hazards arising can be minimised through additional technical-functional tests, which may not yet have been standardised. Risk management is part of a quality management system, which must be bindingly certified for manufacturers and processors of critical medical devices with particularly high processing demands by a body accredited by the competent authority.
Kraft
Marc
M
Technische Universität Berlin, Department of Medical Technology, Berlin, Germany.