title: Control system for the Spallation Neutron Source H- source test facility Allison scanner.
abstract: Spallation Neutron Source is currently in progress of a multiyear plan to ramp ion beam power to the initial design power of 1.4 MW. Key to reaching this goal is understanding and improving the operation of the H(-) ion source. An Allison scanner was installed on the ion source in the test facility to support this improvement. This paper will discuss the hardware and the software control system of the installed Allison scanner. The hardware for the system consists of several parts. The heart of the system is the scanner head, complete with associated bias plates, slits, and signal detector. There are two analog controlled high voltage power supplies to bias the plates in the head, and a motor with associated controller to position the head in the beam. A multifunction data acquisition card reads the signals from the signal detector, as well as supplies the analog voltage control for the power supplies. To synchronize data acquisition with the source, the same timing signal that is used to trigger the source itself is used to trigger data acquisition. Finally, there is an industrial personal computer to control the rest of the hardware. Control software was developed using National Instruments LABVIEW, and consists of two parts: a data acquisition program to control the hardware and a stand alone application for offline user data analysis.
authors.affiliation: Spallation Neutron Source, Oak Ridge National Laboratory, Oak Ridge, Tennessee 37830, USA. l8g@ornl.gov
authors.forename: C D
authors.initials: CD
authors.lastname: Long
authors.forename: M P
authors.initials: MP
authors.lastname: Stockli
authors.forename: T V
authors.initials: TV
authors.lastname: Gorlov
authors.forename: B
authors.initials: B
authors.lastname: Han
authors.forename: S N
authors.initials: SN
authors.lastname: Murray
authors.forename: T R
authors.initials: TR
authors.lastname: Pennisi