Portable emittance measurement device.
In Institute for Theoretical and Experimental Physics (ITEP) the portable emittance measurements device is developed. It provides emittance measurements both with "pepper-pot" and "two slits" methods. Depending on the method of measurements, either slits or pepper-pot mask with scintillator are mounted on the two activators and are installed in two standard Balzer's cross chamber with CF-100 flanges. To match the angle resolution for measured beam, the length of the stainless steel pipe between two crosses changes is adjusted. The description of the device and results of emittance measurements at the ITEP ion source test bench are presented.
Liakin
D
D
Institute for Theoretical and Experimental Physics, Moscow 117218, Russia.
Seleznev
D
D
Orlov
A
A
Kuibeda
R
R
Kropachev
G
G
Kulevoy
T
T
Yakushin
P
P