Preliminary studies for a CBCT imaging protocol for offline organ motion analysis: registration software validation and CTDI measurements.
Cone-beam X-ray volumetric imaging in the treatment room, allows online correction of set-up errors and offline assessment of residual set-up errors and organ motion. In this study the registration algorithm of the X-ray volume imaging software (XVI, Elekta, Crawley, United Kingdom), which manages a commercial cone-beam computed tomography (CBCT)-based positioning system, has been tested using a homemade and an anthropomorphic phantom to: (1) assess its performance in detecting known translational and rotational set-up errors and (2) transfer the transformation matrix of its registrations into a commercial treatment planning system (TPS) for offline organ motion analysis. Furthermore, CBCT dose index has been measured for a particular site (prostate: 120 kV, 1028.8 mAs, approximately 640 frames) using a standard Perspex cylindrical body phantom (diameter 32 cm, length 15 cm) and a 10-cm-long pencil ionization chamber. We have found that known displacements were correctly calculated by the registration software to within 1.3 mm and 0.4°. For the anthropomorphic phantom, only translational displacements have been considered. Both studies have shown errors within the intrinsic uncertainty of our system for translational displacements (estimated as 0.87 mm) and rotational displacements (estimated as 0.22°). The resulting table translations proposed by the system to correct the displacements were also checked with portal images and found to place the isocenter of the plan on the linac isocenter within an error of 1 mm, which is the dimension of the spherical lead marker inserted at the center of the homemade phantom. The registration matrix translated into the TPS image fusion module correctly reproduced the alignment between planning CT scans and CBCT scans. Finally, measurements on the CBCT dose index indicate that CBCT acquisition delivers less dose than conventional CT scans and electronic portal imaging device portals. The registration software was found to be accurate, and its registration matrix can be easily translated into the TPS and a low dose is delivered to the patient during image acquisition. These results can help in designing imaging protocols for offline evaluations.
Falco
Maria Daniela
MD
Department of Diagnostic Imaging, Tor Vergata University General Hospital, Rome, Italy. mdanielafalco@hotmail.com
Fontanarosa
Davide
D
Miceli
Roberto
R
Carosi
Alessandra
A
Santoni
Riccardo
R
D'Andrea
Marco
M
D000465:Algorithms; D016477:Artifacts; D006801:Humans; D009038:Motion; D009862:Online Systems; D019047:Phantoms, Imaging; D010865:Pilot Projects; D011856:Radiographic Image Enhancement / Q000379:methods*; D011857:Radiographic Image Interpretation, Computer-Assisted / Q000379:methods*; D011874:Radiometry / Q000379:methods*; D011879:Radiotherapy Dosage; D015203:Reproducibility of Results; D012680:Sensitivity and Specificity; D012984:Software; D013382:Subtraction Technique; D014057:Tomography, X-Ray Computed / Q000379:methods*