A novel, versatile valved holding chamber for delivering inhaled medications to neonates and small children: laboratory simulation of delivery options.
BACKGROUND
Delivery of bronchodilator to infants and small children from a pressurized metered-dose inhaler with valved holding chamber (pMDI-VHC) is limited by airway narrowness, short respiratory cycle time, and small tidal volume (V(T)). There is a need for a versatile, efficient VHC, given the variety of treatment modalities.


METHODS
We tested the AeroChamber Mini VHC (the internal geometry of which is optimized for aerosol delivery, and which accepts a pMDI canister that has a dose counter) in experiments to determine differences in the delivery of hydrofluoroalkane-propelled albuterol (90 microg/actuation) during: mechanical ventilation via endotracheal tube (ETT); manual resuscitation via ETT; and spontaneous breathing via face mask. We tested 5 units of the AeroChamber Mini VHC per test. We simulated the tidal breathing of a premature neonate (V(T) 6 mL), a term neonate (V(T) 20 mL), and a child approximately 2 years old (V(T) 60 mL). We collected the aerosol on an electret filter and quantitatively assayed for albuterol.


RESULTS
The total emitted mass of albuterol per actuation that exited the VHC was marginally greater during spontaneous breathing (12.1 +/- 1.8 microg) than during manual resuscitation (10.0 +/- 1.1 microg) (P = .046). Albuterol delivery via mechanical ventilation, though comparable with the premature-neonate model (3.3 +/- 1.2 microg), the term-neonate model (3.8 +/- 2.1 microg), and the 2-y-old-child model (4.2 +/- 2.3 microg) (P = .63), was significantly lower than in the spontaneous-breathing and manual-resuscitation models (P < .001). In the neonatal models the total emitted mass was similar with the spontaneous-breathing model (6.0 +/- 1.0 microg with the premature-neonate model, 10.5 +/- 0.7 microg with the term-neonate model) and the manual-resuscitation model (5.5 +/- 0.3 microg premature-neonate model, 10.7 +/- 0.9 microg term-neonate model) (P > or = .46 via one-way analysis of variance).


CONCLUSIONS
The reduced delivery of albuterol during mechanical ventilation (compared to during spontaneous breathing and manual resuscitation via ETT) was probably associated with the saturated atmosphere in the breathing circuit (37 degrees C, relative humidity > 99%), compared to the ambient air (22 +/- 1 degrees C, 44 +/- 7% relative humidity). The AeroChamber Mini VHC may provide a versatile alternative to VHCs that are designed exclusively for one aerosol treatment modality.
DiBlasi
Robert M
RM
Center for Developmental Therapeutics, Seattle Children's Research Institute, Washington, USA.
Coppolo
Dominic P
DP
Nagel
Mark W
MW
Doyle
Cathy C
CC
Avvakoumova
Valentina I
VI
Ali
Rubina S
RS
Mitchell
Jolyon P
JP
D000280:Administration, Inhalation; D000336:Aerosols; D000420:Albuterol / Q000008:administration & dosage*; D001993:Bronchodilator Agents / Q000008:administration & dosage*; D002675:Child, Preschool; D004867:Equipment Design; D006801:Humans; D007223:Infant; D007231:Infant, Newborn; D007234:Infant, Premature; D007442:Intubation, Intratracheal; D008173:Lung Diseases, Obstructive / Q000188:drug therapy*; D008397:Masks; D008954:Models, Biological; D009330:Nebulizers and Vaporizers; D015203:Reproducibility of Results; D012121:Respiration, Artificial
D000336:Aerosols; D001993:Bronchodilator Agents; D000420:Albuterol