Toe gangrene in an infant subsequent to application of adult-type pulse oximeter probe for 10 min.
We present a case of ischemic changes after application of an adult pulse oximeter probe in an infant for a short period of only 10 min. To understand the physiology behind this mishap, we studied pressure exerted by the adult pulse oximeter probe on simulated fingers (fluid-filled pouches), which were filled under gravity through the three-way stopcock. A catheter was placed inside the pouch to continuously measure the pressure. The experiment was performed on different sizes of the finger-shaped pouches and by varying the positioning of the pouches in the pulse oximeter. It was observed that pressure exerted by the pulse oximeter was directly proportional to the extent to which it was stretched after placement on the fluid-filled pouches of different sizes. In the fully extended position, the pressure exerted was higher (up to 30 mmHg) than in the fully collapsed position (4 mmHg). Higher pressures were caused by greater stretch of the probe. This could occur due to larger fingers and closer placement of fingers to the hinge of the probe.
Punj
Jyotsna
J
Department of Anesthesiology, RN 5013, Vth Floor, Academic Section, All India Institute of Medical Sciences, New Delhi, India. Jyotsna_punj@yahoo.com
Jaryal
Ashok
A
Mahalingam
Senthil
S
Mukundan
Chaitra
C
Pandey
Ravindra
R
Darlong
Vanlal
V
Chandralekha
D005734:Gangrene / Q000209:etiology; D006801:Humans; D007223:Infant; D008297:Male; D010092:Oximetry / Q000009:adverse effects*; D011312:Pressure; D014034:Toes / Q000473:pathology*
Crit Care Med. 2000 Oct;28(10):3555-7
11057816
Oral Surg Oral Med Oral Pathol. 1990 Feb;69(2):161-3
2304740
J Appl Physiol. 1976 Apr;40(4):597-600
931880
Anaesthesia. 1997 May;52(5):508-9
9165987
Br J Anaesth. 1992 Aug;69(2):204-5
1389829
Anesth Analg. 2005 Feb;100(2):602
15673915
Pediatr Med Chir. 1995 Jul-Aug;17(4):373-4
7491337
Anesth Analg. 1988 Jul;67(7):712-3
3382049
Zhonghua Yi Xue Za Zhi (Taipei). 1999 Dec;62(12):914-6
10634008