Monte Carlo calculation of monitor unit for electron arc therapy.
OBJECTIVE
Monitor unit (MU) calculations for electron are therapy were carried out using Monte Carlo simulations and verified by measurements. Variations in the dwell factor (DF), source-to-surface distance (SSD), and treatment are angle (a) were studied. Moreover, the possibility of measuring the DF, which requires gantry rotation, using a solid water rectangular, instead of cylindrical, phantom was investigated.


METHODS
A phase space file based on the 9 MeV electron beam with rectangular cutout (physical size = 2.6 x 21 cm2) attached to the block tray holder of a Varian 21 EX linear accelerator (linac) was generated using the EGSnrc-based Monte Carlo code and verified by measurement. The relative output factor (ROF), SSD offset, and DF, needed in the MU calculation, were determined using measurements and Monte Carlo simulations. An ionization chamber, a radiographic film, a solid water rectangular phantom, and a cylindrical phantom made of polystyrene were used in dosimetry measurements.


RESULTS
Percentage deviations of ROF, SSD offset, and DF between measured and Monte Carlo results were 1.2%, 0.18%, and 1.5%, respectively. It was found that the DF decreased with an increase in a, and such a decrease in DF was more significant in the a range of 0 degrees-60 degrees than 60 degrees-120 degrees. Moreover, for a fixed a, the DF increased with an increase in SSD. Comparing the DF determined using the rectangular and cylindrical phantom through measurements and Monte Carlo simulations, it was found that the DF determined by the rectangular phantom agreed well with that by the cylindrical one within +/- 1.2%. It shows that a simple setup of a solid water rectangular phantom was sufficient to replace the cylindrical phantom using our specific cutout to determine the DF associated with the electron arc.


CONCLUSIONS
By verifying using dosimetry measurements, Monte Carlo simulations proved to be an alternative way to perform MU calculations effectively for electron are therapy. Since Monte Carlo simulations can generate a precalculated database of ROF, SSD offset, and DF for the MU calculation, with a reduction in human effort and linac beam-on time, it is recommended that Monte Carlo simulations be partially or completely integrated into the commissioning of electron are therapy.
Chow
James C L
JC
Radiation Medicine Program, Princess Margaret Hospital, University Health Network, Toronto, Ontario M5G 2M9, Canada. james.chow@rmp.uhn.on.ca
Jiang
Runqing
R
D000465:Algorithms; D003198:Computer Simulation; D004583:Electrons; D004867:Equipment Design; D005373:Film Dosimetry / Q000379:methods; D006801:Humans; D009010:Monte Carlo Method; D010315:Particle Accelerators / Q000295:instrumentation; D019047:Phantoms, Imaging; D011137:Polystyrenes / Q000737:chemistry; D011874:Radiometry / Q000379:methods; D011878:Radiotherapy / Q000295:instrumentation* / Q000379:methods*; D011879:Radiotherapy Dosage; D015203:Reproducibility of Results
D011137:Polystyrenes