Effect of semen preparation on casa motility results in cryopreserved bull spermatozoa.
Computer-assisted sperm analyzers (CASA) have become the standard tool for evaluating sperm motility and kinetic patterns because they provide objective data for thousands of sperm tracks. However, these devices are not ready-to-use and standardization of analytical practices is a fundamental requirement. In this study, we evaluated the effects of some settings, such as frame rate and frames per field, chamber and time of analysis, and samples preparations, including thawing temperature, sperm sample concentration, and media used for dilution, on the kinetic results of bovine frozen-thawed semen using a CASA. In Experiment 1, the frame rate (30-60 frame/s) significantly affected motility parameters, whereas the number of frames per field (30 or 45) did not seem to affect sperm kinetics. In Experiment 2, the thawing protocol affects sperm motility and kinetic parameters. Sperm sample concentration significantly limited the opportunity to perform the analysis and the kinetic results. A concentration of 100 and 50 x 10(6) sperm/mL limited the device's ability to perform the analysis or gave wrong results, whereas 5, 10, 20, and 30 x 10(6) sperm/mL concentrations allowed the analysis to be performed, but with different results (Experiment 3). The medium used for the dilution of the sample, which is fundamental for a correct sperm head detection, affects sperm motility results (Experiment 4). In this study, Makler and Leja chambers were used to perform the semen analysis with CASA devices. The chamber used significantly affected motility results (Experiment 5). The time between chamber loading and analysis affected sperm velocities, regardless of chamber used. Based on results recorded in this study, we propose that the CASA evaluation of motility of bovine frozen-thawed semen using Hamilton-Thorne IVOS 12.3 should be performed using a frame rate of 60 frame/s and 30 frames per field. Semen should be diluted at least at 20 x 10(6) sperm/mL using PBS. Furthermore, it is necessary to consider the type of chamber used and perform the analysis within 1 or 2 min, regardless of the chamber used.
Contri
Alberto
A
Department of Veterinary Clinical Sciences, University of Teramo, Viale Crispi 212, 64100 Teramo, Italy. acontri@unite.it
Valorz
Claudio
C
Faustini
Massimo
M
Wegher
Laura
L
Carluccio
Augusto
A
D000818:Animals; D002417:Cattle; D018929:Cell Culture Techniques; D015925:Cryopreservation; D008297:Male; D055101:Semen Analysis / Q000379:methods*; D012662:Semen Preservation; D013081:Sperm Motility; D013094:Spermatozoa / Q000502:physiology*