Quality control and patient dosimetry in dental cone beam CT.
This paper presents the initial experience in performing quality control and patient dose measurements in a cone beam computed tomography (CT) scanner (ILUMA Ultra, IMTEC Imaging, USA) for oral and maxillofacial radiology. The X-ray tube and the generator were tested first, including the kVp accuracy and precision, and the half-value layer (HVL). The following tests specific for panoramic dental systems were also performed: tube output, beam size and beam alignment to the detector. The tests specific for CT included measurements of noise and CT numbers in water and in air, as well as the homogeneity of CT numbers. The most appropriate dose quantity was found to be the air kerma-area product (KAP) measured with a KAP-metre installed at the tube exit. KAP values were found to vary from 110 to 185 microGy m(2) for available adult protocols and to be 54 microGy m(2) for the paediatric protocol. The effective dose calculated with the software PCXMC (STUK, Finland) was 0.05 mSv for children and 0.09-0.16 mSv for adults.
Vassileva
J
J
Department for Radiation Protection at Medical Exposure, National Centre of Radiobiology and Radiation Protection, 132 Kliment Ohridsky blv, 1756 Sofia, Bulgaria. j.vassileva@ncrrp.org
Stoyanov
D
D
D000328:Adult; D001822:Body Burden; D002648:Child; D054893:Cone-Beam Computed Tomography / Q000379:methods*; D006801:Humans; D011785:Quality Assurance, Health Care / Q000379:methods*; D011829:Radiation Dosage; D011835:Radiation Protection / Q000379:methods*; D011861:Radiography, Dental / Q000379:methods*; D011874:Radiometry / Q000379:methods*