The impact of arm position and pulse pressure on the validation of a wrist-cuff blood pressure measurement device in a high risk population.
Despite the increasing popularity of blood pressure (BP) wrist monitors for self-BP measurement at home, device validation and the effect of arm position remains an issue. This study focused on the validation of the Omron HEM-609 wrist BP device, including an evaluation of the impact of arm position and pulse pressure on BP measurement validation. Fifty patients at high risk for cardiovascular disease were selected (age 65 +/- 10 years). Each patient had two measurements with a mercury sphygmomanometer and three measurements with the wrist BP device (wrist at the heart level while the horizontal arm supported [HORIZONTAL], hand supported on the opposite shoulder [SHOULDER], and elbow placed on a desk [DESK]), in random order. The achieved systolic BP (SBP) and diastolic BP (DBP) wrist-cuff readings were compared to the mercury device and the frequencies of the readings within 5, 10, and 15 mmHg of the gold standard were computed and compared with the British Hypertension Society (BHS) and Association for the Advancement of Medical Instrumentation (AAMI) protocols. The results showed while SBP readings with HORIZONTAL and SHOULDER positions were significantly different from the mercury device (mean difference = 7.1 and 13.3 mmHg, respectively; P < 0.05), the DESK position created the closest reading to mercury (mean difference = 3.8, P > 0.1). Approximately 71% of SBP readings with the DESK position were within +/-10 mmHg, whereas it was 62.5% and 34% for HORIZONTAL and SHOULDER positions, respectively. Wrist DBP attained category D with BHS criteria with all three arm positions. Bland-Altman plots illustrated that the wrist monitor systematically underestimated SBP and DBP values. However a reading adjustment of 5 and 10 mmHg for SBP and DBP (DESK position) resulted in improvement with 75% and 77% of the readings being within 10 mmHg (grade B), respectively. AAMI criteria were not fulfilled due to heterogeneity. The findings also showed that the mismatch between the mercury and wrist-cuff systolic BP readings was directly associated with pulse pressure. In conclusion the DESK position produces the most accurate readings when compared to the mercury device. Although wrist BP measurement may underestimate BP measured compared to a mercury device, an adjustment by 5 and 10 mmHg for SBP and DBP, respectively, creates a valid result with the DESK position. Nevertheless, considering the observed variations and the possible impact of arterial stiffness, individual clinical validation is recommended.
Khoshdel
Ali Reza
AR
Faculty of Medicine, Aja University of Medical Science, Tehran, Iran;
Carney
Shane
S
Gillies
Alastair
A
BMJ. 2001 Mar 3;322(7285):531-6
11230071
Blood Press Monit. 2000 Jun;5(3):175-80
10915231
Blood Press Monit. 2002 Oct;7(5):281-4
12409888
Arq Bras Cardiol. 2001 Dec;77(6):532-40
11799428
Blood Press Monit. 2002 Feb;7(1):3-17
12040236
Blood Press. 2002;11(6):357-65
12523679
J Hypertens. 1998 Mar;16(3):259-61
9557917
G Ital Nefrol. 2003 Nov-Dec;20(6):589-91
14732910
Public Health. 2000 May;114(3):165-8
10878742
Hypertension. 2001 May;37(5):1236-41
11358934
Blood Press Monit. 2004 Apr;9(2):77-81
15096904
Blood Press Monit. 2005 Apr;10(2):67-71
15812253
Hypertens Res. 2003 Oct;26(10):771-82
14621179
J Hypertens. 2002 Apr;20(4):573-8
11910283
Anaesth Intensive Care. 2002 Feb;30(1):43-7
11939439
Blood Press Monit. 2004 Feb;9(1):31-7
15021076
Am J Hypertens. 1998 Dec;11(12):1469-74
9880129
Blood Press Monit. 2002 Aug;7(4):237-41
12198340
J Hypertens. 2002 Apr;20(4):629-38
11910297
BMJ. 2001 May 12;322(7295):1167-70
11348915
IEEE Trans Biomed Eng. 2002 Sep;49(9):1038-44
12214876