title: Influence of axial movement on fatigue of ProFile Ni-Ti rotary instruments: an in vitro evaluation.
abstract: The aim of this study was to evaluate the influence of the axial movement and the angle of curve (in degrees) on fatigue of nickel-titanium (Ni-Ti) ProFile rotary endodontic instruments. Ni-Ti ProFile rotary instruments (Maillefer SA, Ballaigues, Switzerland), 25 mm long in the range of ISO size 15 to 40 with two tapers (0.4 and 0.6) were evaluated. They are divided in two groups: the instruments with axial movement and those without axial movement. The system used to test the fatigue is maintained in mechanical conditions as close as possible to the clinical situation. The axial movement is in the order of 2 mm in corono-apical direction with a frequency of 1 Hz. The concave radii incorporating a notched V-form for guiding the instruments were: 5; 7,5 and 10 mm. The rotary system is mounted on an electric handpiece and rotated at 350 rpm speed as recommended by the manufacturers. The instruments are rotated until their separation, and the time, in seconds, is recorded. Statistical evaluation is undertaken using a two-way t-test to identify significant differences between variables in the study (p <0.05). We found significant statistical difference (p<0.05) between Ni-Ti engine drive ProFile instruments incorporating an axial movement and the instruments without axial movement with the same radius of curvature, size and taper.The incorporation of the axial movement increases significantly the life-span of the ProFile rotary instruments. This should reduce the risk of the instrument separation during the endodontic treatment.
authors.affiliation: INSERM Unit 977, 11 Rue Humann, Strasbourg Cedex, France.
authors.forename: Marie-Chantal
authors.initials: MC
authors.lastname: Avoaka
authors.forename: Youssef
authors.initials: Y
authors.lastname: Haïkel
mesh_terms: D003759:Dental Implantation, Endosseous, Endodontic / Q000295:instrumentation*; D004868:Equipment Failure; D008422:Materials Testing; D009068:Movement; D009532:Nickel; D009967:Orthodontic Appliances; D013314:Stress, Mechanical; D014025:Titanium
chemical_list: C040654:titanium nickelide; D009532:Nickel; D014025:Titanium