Coronary guidewire circumcision during use of a Gopher support catheter: potential adverse interaction with polymer-jacketed wire design.
Support catheters with a threaded, distal tip may be useful for crossing recalcitrant coronary lesions. The interaction between the Gopher support catheter and polymer-coated coronary guidewires may, however, result in circumferential stripping of polymer from the guidewire, with resulting wire damage or fracture. We report two cases of polymer-coated guidewire damage with use of a Gopher support catheter, suggesting that these two devices should not be used together.
Armstrong
Ehrin J
EJ
Department of Medicine, Division of Cardiology, University of California-San Francisco, CA, USA.
Shunk
Kendrick A
KA
D000368:Aged; D015906:Angioplasty, Balloon, Coronary / Q000295:instrumentation*; D017225:Atherectomy, Coronary; D002908:Chronic Disease; D020099:Coated Materials, Biocompatible; D017023:Coronary Angiography; D054059:Coronary Occlusion / Q000000981:diagnostic imaging / Q000628:therapy*; D023921:Coronary Stenosis / Q000000981:diagnostic imaging / Q000628:therapy*; D020878:Device Removal; D004867:Equipment Design; D004868:Equipment Failure; D006801:Humans; D008297:Male; D008875:Middle Aged; D012720:Severity of Illness Index; D016896:Treatment Outcome
D020099:Coated Materials, Biocompatible