title: IMRT implementation and patient specific dose verification with film and ion chamber array detectors.
abstract: Implementation of Intensity Modulation Radiotherapy (IMRT) and patient dose verification was carried out with film and I'mariXX using linear accelerator with 120-leaf Millennium dynamic multileaf collimator (dMLC). The basic mechanical and electrical commissioning and quality assurance tests of linear accelerator were carried out. The leaf position accuracy and leaf position repeatability checks were performed for static MLC positions. Picket fence test and garden fence test were performed to check the stability of the dMLC and the reproducibility of the gap between leaves. The radiation checks were performed to verify the position accuracy of MLCs in the collimator system. The dMLC dosimetric checks like output stability, average leaf transmission and dosimetric leaf separation were also investigated. The variation of output with gravitation at different gantry angles was found to be within 0.9 %. The measured average leaf transmission for 6 MV was 1.6 % and 1.8% for 18 MV beam. The dosimetric leaf separation was found to be 2.2 mm and 2.3 mm for 6 MV and 18 MV beams. In order to check the consistency of the stability and the precision of the dMLC, it is necessary to carryout regular weekly and monthly checks. The dynalog files analysis for Garden fence, leaf gap width and step wedge test patterns carried out weekly were in good agreement. Pretreatment verification was performed for 50 patients with ion chamber and I'mariXX device. The variations of calculated absolute dose for all treatment fields with the ion chamber measurement were within the acceptable criterion. Treatment Planning System (TPS) calculated dose distribution pattern was comparable with the I'mariXX measured dose distribution pattern. Out of 50 patients for which the comparison was made, 36 patients were agreed with the gamma pixel match of >95% and 14 patients were with the gamma pixel match of 90-95% with the criteria of 3% delta dose (DD) and 3 mm distance-to-agreement (DTA). Commissioning and quality assurance of dMLC for IMRT application requires considerable time and effort. Many dosimetric characteristics need to be assessed carefully failing which the delivered dose will be significantly different from the planned dose. In addition to the issues discussed above we feel that individual MU check is necessary before the treatment is delivered.
authors.affiliation: Department of Radiation Physics, Kidwai Memorial Institute of Oncology, Bangalore, India. ssathiyan@rediffmail.com
authors.forename: S
authors.initials: S
authors.lastname: Sathiyan
authors.forename: M
authors.initials: M
authors.lastname: Ravikumar
authors.forename: C
authors.initials: C
authors.lastname: Varatharaj
authors.forename: S S
authors.initials: SS
authors.lastname: Supe
authors.forename: S L
authors.initials: SL
authors.lastname: Keshava
mesh_terms: D005373:Film Dosimetry / Q000295:instrumentation*; D006801:Humans; D011879:Radiotherapy Dosage; D011880:Radiotherapy Planning, Computer-Assisted / Q000295:instrumentation; D050397:Radiotherapy, Intensity-Modulated / Q000295:instrumentation*