Measured Temperature Dependence of Scintillation Camera Signals Read Out by Geiger-Müller Mode Avalanche Photodiodes.
We are developing a prototype monolithic scintillation camera with optical sensors on the entrance surface (SES) for use with statistically-estimated depth-of-interaction in a continuous scintillator. We opt to use Geiger-Müller mode avalanche photodiodes (GM-APDs) for the SES camera since they possess many desirable properties; for the intended application (SES and PET/MR imaging), they offer a thin attenuation profile and an operational insensitivity to large magnetic fields. However, one issue that must be addressed in using GM-APDs in an RF environment (as in MR scanners) is the thermal dissipation that can occur in this semiconductor material.Signals of GM-APDs are strongly dependent on junction temperature. Consequently, we are developing a temperature-controlled GM-APD-based PET camera whose monitored temperature can be used to dynamically account for the temperature dependence of the output signals. Presently, we aim to characterize the output-signal dependence on temperature and bias for a GM-APD-based scintillation camera.We've examined two GM-APDs, a Zecotek prototype MAPD-3N, and a SensL commercial SPMArray2. The dominant effect of temperature on gain that we observe results from a linear dependence of breakdown voltage on temperature (0.071 V/°C and 0.024 V/°C, respectively); at 2.3 V excess bias (voltage above breakdown) the resulting change in gain with temperature (without adjusting bias voltage) is -8.5% per °C for the MAPD-3N and -1.5 % per °C for the SPMArray2. For fixed excess bias, change in dark current with temperature varied widely, decreasing by 25% to 40% as temperature was changed from 20 °C to 10 °C and again by 20% to 35% going from 10 °C to 0 °C. Finally, using two MAPD-3N to read out a pair of 3.5-by-3.5-by-20 mm(3) Zecotek LFS-3 scintillators in coincidence, we observe a decrease from 1.7 nsec to 1.5 nsec in coincidence-time resolution as we lowered temperature from 23 °C to 10 °C.
Hunter
William C J
WC
IEEE.
Miyaoka
Robert S
RS
Macdonald
L R
LR
Lewellen
Thomas K
TK