A quick diagnostic test for NMR receiver gain compression.
Modern NMR spectrometers require receivers to work within their linear ranges to maintain high fidelity line shape and peak integration. For better sensitivity, the receiver gain has to be optimized to detect dilute analytes; however, gain compression needs to be avoided. Here, we explore if and how linear receiver performance can be achieved for a couple of representative gain settings on a spectrometer. In the case of slight receiver gain compression, not only will the peak integral be attenuated but a very small line-shape change can also be observed. Hence, we can resort to resonance integration and line-shape analysis for gain compression diagnosis. As such, NMR signals, regardless of their observed amplitude difference in frequency domain, can be accurately compared in quantitative analysis.
Mo
Huaping
H
Purdue Interdepartmental NMR Facility, Purdue University, West Lafayette, IN 47907, USA. hmo@purdue.edu
Harwood
John S
JS
Raftery
Daniel
D
D044962:Data Compression; D009682:Magnetic Resonance Spectroscopy / Q000379:methods*; D012815:Signal Processing, Computer-Assisted
J Nat Prod. 2009 Apr;72(4):739-44
19399996
Anal Chem. 1999 Jul 1;71(13):2554-7
21662801
Phys Rev B Condens Matter. 1994 Jun 1;49(22):15702-15711
10010702
Nat Protoc. 2007;2(11):2692-703
18007604
Anal Bioanal Chem. 2004 Mar;378(6):1568-73
15214418
Magn Reson Chem. 2010 Mar;48(3):235-8
20063326
J Magn Reson. 2009 Oct;200(2):239-44
19647457
Anal Chem. 2008 Dec 15;80(24):9835-9
19007190
J Nat Prod. 2008 May;71(5):810-3
18393462