title: Acetabular UHMWPE survival and wear changes with different manufacturing techniques.
abstract: BACKGROUND
  Polyethylene wear may be affected by the type of polyethylene resin, manufacturing technique, degree of thermal stabilization, and sterilization technique.
  
  
  OBJECTIVE
  We therefore compared femoral head penetration into the PE and cup survival using the same cup system with different PE resins, manufacturing, and sterilization techniques.
  
  
  METHODS
  Our study group consisted of 1912 THAs performed using the same uncemented cup and identical 28-mm cobalt-chrome heads. The polyethylene varied as follows: Group 1 (94 cups), GUR 4150 resin, ram-extruded, sterilized in air, no barrier packaging; Group 2 (74 cups), same as Group 1 but sterilized in argon; Group 3 (75 cups), Himont 1900 resin, compression-molded bar stock, sterilized in argon, no barrier packaging; Group 4 (620 cups), same as Group 3 except with barrier packing; Group 5 (711 cups), GUR 1050 resin, compression-molded bar stock, sterilized in argon gas with barrier packaging; and Group 6 (338 cups), GUR 1050 resin, compression-molded bar stock, sterilized in argon with barrier packaging, irradiated with 50 kGy, heated below melting temperature, machined, and finally placed in nonbarrier packaging with gas plasma sterilization. Minimum followup was 2 years (average, 7 years; range, 2-17 years).
  
  
  RESULTS
  Femoral head penetration averaged 0.05 mm per year for Groups 5 and 6 and was substantially lower than for Groups 1 to 4. Cup survival was higher at seven years in Groups 3, 4, and 5, and at 10 years in group 4 when compared to groups 1, 2, and 3.
  
  
  CONCLUSIONS
  We observed lower FHP rates and higher cup survival with polyethylene machined from direct compression-molded bar stock, sterilized in argon gas, with barrier packaging.
  
  
  METHODS
  Level III Therapeutic study. See Guidelines for Authors for a complete description of levels of evidence.
authors.affiliation: The Center for Hip and Knee Surgery, St Francis Hospital, Mooresville, 1199 Hadley Road, Mooresville, IN 46158, USA. jmeding@msn.com
authors.forename: John B
authors.initials: JB
authors.lastname: Meding
authors.forename: E Michael
authors.initials: EM
authors.lastname: Keating
authors.forename: Kenneth E
authors.initials: KE
authors.lastname: Davis
mesh_terms: D000077:Acetabulum / Q000601:surgery*; D000293:Adolescent; D000328:Adult; D000368:Aged; D000369:Aged, 80 and over; D019644:Arthroplasty, Replacement, Hip / Q000295:instrumentation* / Q000379:methods; D001672:Biocompatible Materials; D001696:Biomechanical Phenomena; D005260:Female; D005270:Femur Head / Q000000981:diagnostic imaging / Q000473:pathology; D006622:Hip Prosthesis; D006801:Humans; D007592:Joint Diseases / Q000601:surgery; D008297:Male; D008422:Materials Testing / Q000379:methods*; D008875:Middle Aged; D011095:Polyethylenes; D011475:Prosthesis Failure; D011859:Radiography; D013499:Surface Properties; D055815:Young Adult
chemical_list: D001672:Biocompatible Materials; D011095:Polyethylenes; C111601:ultra-high molecular weight polyethylene