A new certified reference material for benzene measurement in air on a sorbent tube: development and proficiency testing.
A certified matrix reference material (CRM) for the measurement of benzene in ambient air has been developed at Laboratoire National de Métrologie et d'Essais. The production of these CRMs was conducted using a gravimetric method fully traceable to the International System of Units. The CRMs were prepared by sampling an accurate mass of a gaseous primary reference material of benzene, using a high-precision laminar flowmeter and a mass flow controller, with a PerkinElmer sampler filled with Carbopack™ X sorbent. The relative standard deviations obtained for the preparation of a batch of 20 tubes loaded with 500 ng of benzene were below 0.2%. Each CRM is considered independent from the others and with its own certified value and an expanded uncertainty estimated to be within 0.5%, lower than the uncertainties of benzene CRMs already available worldwide. The stability of these materials was also established up to 12 months. These CRMs were implemented during proficiency testing, to evaluate the analytical performances of seven French laboratories involved in benzene air monitoring.
Caurant
A
A
Laboratoire National de Métrologie et d'Essais, 1 rue Gaston Boissier, 75724 Paris, Cedex 15, France. adrien.caurant@lne.fr
Lalère
B
B
Schbath
M-C
MC
Stumpf
C
C
Sutour
C
C
Mace
T
T
Quisefit
J-P
JP
Doussin
J-F
JF
Vaslin-Reimann
S
S
D000042:Absorption; D000393:Air Pollutants / Q000032:analysis*; D001554:Benzene / Q000032:analysis*; D002849:Chromatography, Gas; D004784:Environmental Monitoring / Q000379:methods* / Q000592:standards*; D012015:Reference Standards
D000393:Air Pollutants; D001554:Benzene