Electret ion chamber-based passive radon-thoron discriminative monitors.
Electret ion chambers (EICs), commercially available under brand name E-PERM(®), are widely used for measuring indoor and outdoor (222)Rn concentrations in air. These are designed to respond only to (222)Rn and not to (220)Rn by restricting diffusional entry area. Such radon EIC (R EIC) monitors are modified by increasing the entry area to allow (220)Rn, in addition to (222)Rn. Such modified units are called RT EIC. When a set of R and RT EICs are collocated, it is possible to discriminate and measure both radon and thoron concentrations, using appropriate calibration factors (CFs) and algorithms. The EICs come in different volumes, providing different sensitivities. The thoron CFs for 58-, 210- and 960-ml volume R and RT pairs are, respectively, 2.8-, 18.7- and 89-V drop per (kBq m(-3) d ), respectively. These provide much wider sensitivities and ranges compared to alpha track-based passive radon-thoron discriminative monitors.
Kotrappa
P
P
Rad Elec Inc., 5716-A Industry Lane, Frederick, MD 21704, USA. pkotrappa@aol.com
Steck
D
D
D000396:Air Pollutants, Radioactive; D016902:Air Pollution, Indoor / Q000032:analysis*; D000465:Algorithms; D002138:Calibration; D004867:Equipment Design; D007477:Ions; D015233:Models, Statistical; D011834:Radiation Monitoring / Q000295:instrumentation*; D011874:Radiometry; D011886:Radon / Q000032:analysis*; D017687:Radon Daughters / Q000032:analysis*; D012306:Risk; D013997:Time Factors
D000396:Air Pollutants, Radioactive; D007477:Ions; D017687:Radon Daughters; D011886:Radon