title: Disinfectant wipes containing hydrogen peroxide induce overestimation of glucose results obtained with Lifescan SureStep Flexx® glucose meter.
abstract: OBJECTIVE
  Investigate the interference of Virox® disinfectant wipes containing hydrogen peroxide with Lifescan SureStep Flexx® glucose meter results.
  
  
  METHODS
  Glucose measurements of control and whole blood samples were performed before and after cleaning with Virox® wipes to reveal the kinetics of this interference.
  
  
  CONCLUSIONS
  After a period where the error code 2(b) is given by the device, an overestimation period of glucose results is always present at all levels tested for a duration of hours. As this interference cannot be detected by the operator, hydrogen peroxide containing wipes for sanitizing of the glucose oxidase based glucose meter must be forbidden.
authors.affiliation: Department of Clinical Biochemistry, CHU Ste-Justine, Université de Montréal, Montréal, QC, Canada. philippe.desmeules.hse@ssss.gouv.qc.ca
authors.forename: Philippe
authors.initials: P
authors.lastname: Desmeules
authors.forename: Johanne
authors.initials: J
authors.lastname: Ethier
authors.forename: Pierre
authors.initials: P
authors.lastname: Allard
mesh_terms: D001786:Blood Glucose / Q000032:analysis*; D004202:Disinfectants / Q000009:adverse effects*; D004864:Equipment and Supplies / Q000592:standards*; D006801:Humans; D006861:Hydrogen Peroxide / Q000009:adverse effects*; D008991:Monitoring, Physiologic / Q000295:instrumentation*; D019095:Point-of-Care Systems
chemical_list: D001786:Blood Glucose; D004202:Disinfectants; D006861:Hydrogen Peroxide