Comparison of (semi-)automatic and manually adjusted measurements of left ventricular function in dual source computed tomography using three different software tools.
To assess the accuracy of (semi-)automatic measurements of left ventricular (LV) functional parameters in cardiac dual-source computed tomography (DSCT) compared to manually adjusted measurements in three different workstations. Forty patients, who underwent cardiac DSCT, were included (31 men, mean age 58 ± 14 years). Multiphase reconstructions were made with ten series at every 10% of the RR-interval. LV function analysis was performed on three different, commercially available workstations. On all three workstations, end-systolic volume (ESV), end-diastolic volume (EDV), LV ejection fraction (LVEF) and myocardial mass (MM) were calculated as automatically as possible. With the same DSCT datasets, LV functional parameters were also calculated with as many manual adjustments as needed for accurate assessment for all three software tools. For both semi-automatic as well as manual methods, time needed for evaluation was recorded. Paired t-tests were employed to calculate differences in LV functional parameters. Repeated measurements were performed to determine intra-observer and inter-observer variability. (Semi-)automatic measurements revealed a good correlation with manually adjusted measurements for Vitrea (LVEF r = 0.93, EDV r = 0.94, ESV r = 0.98 and MM r = 0.94) and Aquarius (LVEF r = 0.96, EDV r = 0.94, ESV r = 0.98 and MM r = 0.96). Also, good correlation was obtained for Circulation, except for LVEF (LVEF r = 0.45, EDV r = 0.93, ESV r = 0.92 and MM r = 0.86). However, statistically significant differences were found between (semi-)automatically and manually adjusted measurements for LVEF (P < 0.05) and ESV (P < 0.001) in Vitrea, all LV functional parameters in Circulation (P < 0.001) and EDV, ESV and MM (<0.001) in Aquarius Workstation. (Semi-)automatic measurement of LV functional parameters is feasible, but significant differences were found for at least two different functional parameters in all three workstations. Therefore, expert manual correction is recommended at all times.
Department of Radiology, University Medical Center Groningen, University of Groningen, Hanzeplein 1, P.O. Box 30.001, 9700 RB, Groningen, The Netherlands. g.de.jonge@rad.umcg.nl
G J
GJ
de Jonge
P M A
PM
van Ooijen
J
J
Overbosch
A Litcheva
AL
Gueorguieva
M C
MC
Janssen-van der Weide
M
M
Oudkerk
D000328:Adult; D000368:Aged; D000369:Aged, 80 and over; D057205:Automation, Laboratory; D005260:Female; D006801:Humans; D008297:Male; D008875:Middle Aged; D009426:Netherlands; D015588:Observer Variation; D011237:Predictive Value of Tests; D011857:Radiographic Image Interpretation, Computer-Assisted; D015203:Reproducibility of Results; D012984:Software; D013318:Stroke Volume; D014057:Tomography, X-Ray Computed; D018487:Ventricular Dysfunction, Left / Q000000981:diagnostic imaging* / Q000503:physiopathology; D016277:Ventricular Function, Left