Appropriate patient selection at abdominal dual-energy CT using 80 kV: relationship between patient size, image noise, and image quality.
OBJECTIVE
To determine the computed tomographic (CT) detector configuration, patient size, and image noise limitations that will result in acceptable image quality of 80-kV images obtained at abdominal dual-energy CT.


METHODS
The Institutional Review Board approved this HIPAA-compliant retrospective study from archival material from patients consenting to the use of medical records for research purposes. A retrospective review of contrast material-enhanced abdominal dual-energy CT scans in 116 consecutive patients was performed. Three gastrointestinal radiologists noted detector configuration and graded image quality and artifacts at specified levels-midliver, midpancreas, midkidneys, and terminal ileum-by using two five-point scales. In addition, an organ-specific enhancement-to-noise ratio and background noise were measured in each patient. Patient size was measured by using the longest linear dimension at the level of interest, weight, lean body weight, body mass index, and body surface area. Detector configuration, patient sizes, and image noise levels that resulted in unacceptable image quality and artifact rankings (score of 4 or higher) were determined by using multivariate logistic regression.


RESULTS
A 14 × 1.2-mm detector configuration resulted in fewer images with unacceptable quality than did the 64 × 0.6-mm configuration at all anatomic levels (P = .004, .01, and .02 for liver, pancreas, and kidneys, respectively). Image acceptability for the kidneys and ileum was significantly greater than that for the liver for all readers and detector configurations (P < .001). For the 14 × 1.2-mm detector configuration, patient longest linear dimensions yielding acceptable image quality across readers ranged from 34.9 to 35.8 cm at the four anatomic levels.


CONCLUSIONS
An 80-kV abdominal CT can be performed with appropriate diagnostic quality in a substantial percentage of the population, but it is not recommended beyond the described patient size for each anatomic level. The 14 × 1.2-mm detector configuration should be preferred.
Department of Radiology, Mayo Clinic, 200 First St SW, Rochester, MN 55905, USA.
Luís S
LS
Guimarães
Joel G
JG
Fletcher
William S
WS
Harmsen
Lifeng
L
Yu
Hassan
H
Siddiki
Zachary
Z
Melton
James E
JE
Huprich
David
D
Hough
Robert
R
Hartman
Cynthia H
CH
McCollough
D000328:Adult; D000368:Aged; D000369:Aged, 80 and over; D016009:Chi-Square Distribution; D003287:Contrast Media; D005260:Female; D006801:Humans; D016015:Logistic Models; D008297:Male; D008875:Middle Aged; D018579:Patient Selection; D011786:Quality Control; D011860:Radiography, Abdominal / Q000379:methods* / Q000592:standards; D012189:Retrospective Studies; D018709:Statistics, Nonparametric; D014057:Tomography, X-Ray Computed / Q000379:methods* / Q000592:standards
D003287:Contrast Media