title: Field safety notices released by manufacturers in cases of failure of products for infection testing: analysis of cases reported to the BfArM between 2005 and 2007.
abstract: The European Directive 98/79/EC for in vitro diagnostic medical devices (IVD) regulates marketing and post marketing surveillance of IVD in the European Economic Area. Manufacturers have to inform the responsible Competent Authorities (CA) about incidents and field safety corrective actions (FSCA) related to IVD. In Germany, the Federal Institute for Drugs and Medical Devices (BfArM) is the responsible CA for most IVD, only few IVD as specified in Annex II of the Directive are under the responsibility of the Paul Ehrlich Institute (PEI). In case of a FSCA manufacturers have to inform customers by means of a Field Safety Notice (FSN) which should be sent to BfArM prior to release and is published on the BfArM homepage. Between beginning of 2005 and end of 2007 the BfArM received a total of 1025 reports regarding IVD. From these, 38 related to tests, reagents, calibrators, and control materials for infection testing, 13 related to analysers and general consumables (n = 8 and n = 5, respectively) based on culture techniques, and 7 related to analysers and general consumables (n = 5 and n = 2, respectively) based on molecular biological methods. FSCA were performed in Germany in 32 (84.2%) of all notifications related to tests reagents, calibrators, and control materials as well as in 13 (100%) and 7 (100%) of notifications related to analysers and consumables based on culture techniques and molecular biological methods, respectively. A number of relevant deficiencies regarding the quality of the FSN were separately demonstrated for FSN in German and English language. In brief, manufacturers often sent their FSN to the BfArM with delay. Additionally, a subset of FSN provided insufficient information on the product related risks or the measures to be performed by the customer to mitigate product related risks. Furthermore, customer confirmation forms often were missing in the FSN sent to the BfArM. Our data suggest that for IVD for infection testing FSCA and FSN are frequently performed. For better vigilance performance, manufacturers could shorten the time until release and improve the contents of FSN to ensure the safety of IVD in cases of product related corrective actions.
authors.affiliation: Federal Institute for Drugs and Medical Devices, 53175 Bonn, Germany. r.siekmeier@bfarm.de
authors.forename: R
authors.initials: R
authors.lastname: Siekmeier
authors.forename: K
authors.initials: K
authors.lastname: Lisson
authors.forename: D
authors.initials: D
authors.lastname: Wetzel
mesh_terms: D002138:Calibration; D004868:Equipment Failure; D004864:Equipment and Supplies; D006801:Humans; D007239:Infections / Q000175:diagnosis*; D011358:Product Surveillance, Postmarketing; D013997:Time Factors