title: Investigation of a Multi-Anode Microchannel Plate PMT for Time-of-Flight PET.
abstract: We report on an investigation of a mulit-anode microchannel plate PMT for time-of-flight PET detector modules. The primary advantages of an MCP lie in its excellent timing properties (fast rise time and low transit time spread), compact size, and reasonably large active area, thus making it a good candidate for TOF applications. In addition, the anode can be segmented into an array of collection electrodes with fine pitch to attain good position sensitivity. In this paper, we investigate using the Photonis Planacon MCP-PMT with a pore size of 10 µm to construct a PET detector module, specifically for time-of-flight applications. We measure the single electron response by exciting the Planacon with pulsed laser diode. We also measure the performance of the Planacon as a PET detector by coupling a 4 mm × 4 mm × 10 mm LSO crystal to individual pixel to study its gain uniformity, energy resolution, and timing resolution. The rise time of the Planacon is 440 ps with pulse duration of about 1 ns. A transit time spread of 120 ps FWHM is achieved. The gain is fairly uniform across the central region of the Planacon, but drops off by as much as a factor of 2.5 around the edges. The energy resolution is fairly uniform across the Planacon with an average value of 18.6±0.7% FWHM. While the average timing resolution of 252±7 ps FWHM is achieved in the central region of the Planacon, it degrades to 280±9 ps FWHM for edge pixels and 316±15 ps FWHM for corner pixels. We compare the results with measurements performed with a fast timing conventional PMT (Hamamatsu R-9800). We find that the R9800, which has significantly higher PDE, has a better timing resolution than the Planacon. Furthermore, we perform detector simulations to calculate the improvement that can be achieved with a higher PDE Planacon. The calculation shows that the Planacon can achieve significantly better timing resolution if it can attain the same PDE as the R-9800, while only a 30% improvement is needed to yield a similar timing resolution as the R-9800.
authors.affiliation: W.-S. Choong is with the Lawrence Berkeley National Laboratory, Berkeley, CA 94720 USA (phone: 510-486-6757; fax: 510-486-4768;  wschoong@lbl.gov ).
authors.forename: Woon-Seng
authors.initials: WS
authors.lastname: Choong