title: Field trial of three different Plasmodium vivax-detecting rapid diagnostic tests with and without evaporative cool box storage in Afghanistan.
abstract: BACKGROUND
  Accurate parasitological diagnosis of malaria is essential for targeting treatment where more than one species coexist. In this study, three rapid diagnostic tests (RDTs) (AccessBio CareStart (CSPfPan), CareStart PfPv (CSPfPv) and Standard Diagnostics Bioline (SDBPfPv)) were evaluated for their ability to detect natural Plasmodium vivax infections in a basic clinic setting. The potential for locally made evaporative cooling boxes (ECB) to protect the tests from heat damage in high summer temperatures was also investigated.
  
  
  METHODS
  Venous blood was drawn from P. vivax positive patients in Jalalabad, Afghanistan and tested against a panel of six RDTs. The panel comprised two of each test type; one group was stored at room temperature and the other in an ECB. RDT results were evaluated against a consensus gold standard based on two double-read reference slides and PCR. The sensitivity, specificity and a measure of global performance for each test were determined and stratified by parasitaemia level and storage condition.
  
  
  RESULTS
  In total, 306 patients were recruited, of which 284 were positive for P. vivax, one for Plasmodium malariae and none for Plasmodium falciparum; 21 were negative. All three RDTs were specific for malaria. The sensitivity and global performance index for each test were as follows: CSPfPan [98.6%, 95.1%], CSPfPv [91.9%, 90.5%] and SDBPfPv [96.5%, 82.9%], respectively. CSPfPv was 16% less sensitive to a parasitaemia below 5,000/μL. Room temperature storage of SDBPfPv led to a high proportion of invalid results (17%), which reduced to 10% in the ECB. Throughout the testing period, the ECB maintained ~8°C reduction over ambient temperatures and never exceeded 30°C.
  
  
  CONCLUSIONS
  Of the three RDTs, the CSPfPan test was the most consistent and reliable, rendering it appropriate for this P. vivax predominant region. The CSPfPv test proved unsuitable owing to its reduced sensitivity at a parasitaemia below 5,000/μL (affecting 43% of study samples). Although the SDBPfPv device was more sensitive than the CSPfPv test, its invalid rate was unacceptably high. ECB storage reduced the proportion of invalid results for the SDBPfPv test, but surprisingly had no impact on RDT sensitivity at low parasitaemia.
authors.affiliation: London School of Hygiene & Tropical Medicine, London, UK. Amy.Mikhail@lshtm.ac.uk
authors.forename: Amy F W
authors.initials: AF
authors.lastname: Mikhail
authors.forename: Toby J
authors.initials: TJ
authors.lastname: Leslie
authors.forename: Mohammad I
authors.initials: MI
authors.lastname: Mayan
authors.forename: Rohullah
authors.initials: R
authors.lastname: Zekria
authors.forename: Nader
authors.initials: N
authors.lastname: Mohammad
authors.forename: Mohammad A
authors.initials: MA
authors.lastname: Hasanzai
authors.forename: Najibullah
authors.initials: N
authors.lastname: Safi
authors.forename: Christopher J M
authors.initials: CJ
authors.lastname: Whitty
authors.forename: Mark
authors.initials: M
authors.lastname: Rowland
mesh_terms: D000293:Adolescent; D000328:Adult; D000346:Afghanistan; D000368:Aged; D000369:Aged, 80 and over; D002648:Child; D003955:Diagnostic Tests, Routine / Q000379:methods*; D006801:Humans; D016780:Malaria, Vivax / Q000175:diagnosis*; D008875:Middle Aged; D018512:Parasitemia / Q000175:diagnosis*; D010966:Plasmodium vivax / Q000302:isolation & purification*; D012034:Refrigeration / Q000379:methods*; D012680:Sensitivity and Specificity; D013048:Specimen Handling / Q000379:methods*; D055815:Young Adult