title: A practical guide to neonatal volume guarantee ventilation.
abstract: A recent systematic review and meta-analysis shows that volume-targeted ventilation (VTV) compared with pressure-limited ventilation (PLV) reduce death and bronchopulmonary dysplasia, pneumothorax, hypocarbia and severe cranial ultrasound abnormalities. In this paper, we present published research and our experience with volume guarantee (VG) ventilation, a VTV mode available on the Dräger Babylog 8000plus and VN500 ventilators. The VG algorithm measures the expired tidal volume (V(T)) for each inflation and adjusts the peak inflating pressure for the next inflation to deliver a V(T) set by the clinician. The advantage of controlling expired V(T) is that this is less influenced by endotracheal tube leak than inspired V(T). VG ventilation can be used with an endotracheal tube leak up to ∼50%. Initial set V(T) for infants with respiratory distress syndrome should be 4.0 to 5.0 ml kg(-1). The set V(T) should be adjusted to maintain normocapnoea. Setting the peak inflating pressure limit well above the working pressure is important to enable the ventilator to deliver the set V(T), and to avoid frequent alarms. This paper provides a practical guide on how to use VG ventilation.
authors.affiliation: Department of Paediatrics, University Hospital of North Norway, Tromsø, Norway. claus.klingenberg@unn.no
authors.forename: C
authors.initials: C
authors.lastname: Klingenberg
authors.forename: K I
authors.initials: KI
authors.lastname: Wheeler
authors.forename: P G
authors.initials: PG
authors.lastname: Davis
authors.forename: C J
authors.initials: CJ
authors.lastname: Morley
mesh_terms: D006801:Humans; D007231:Infant, Newborn; D017410:Practice Guidelines as Topic; D012121:Respiration, Artificial / Q000295:instrumentation / Q000379:methods*; D012127:Respiratory Distress Syndrome, Newborn / Q000628:therapy*; D012122:Ventilators, Mechanical