title: [Computerized temperature monitoring of the vaccine cold chain in a tropical climate, Chad].
abstract: Because new EPI liquid vaccines are highly sensitive to freezing and overheating, close monitoring of the cold chain is mandatory. The new Testostore 171-1 electronic thermometer (Testo) provides more reliable monitoring of cold chain temperature than freezer indicators, vaccine vial monitors and color strips that only indicate if vaccines are out-of-date. The Testo thermometer uses a probe placed in refrigeration units to periodically measure and store temperature readings. Temperature curves are displayed via a USB connection on a laptop computer running special software (Comfort software light). Testo temperature data can easily be communicated to all management levels by e-mail. The first experience using the Testo system in Africa involved regional EPI supervision in Mondou, Logone Occidental, Chad. After a preliminary mission in Chad in 2006 showed the feasibility of using this method to manage the national cold chain at all levels, a nurse was appointed as EPI supervisor and given a refresher course in Chad's capital Ndjamena in March 2009. In April-May 2009, the supervisor was sent back to the Logone Occidental Region to monitor, by himself, refrigeration units making up the regional and district cold chain for vaccine storage in five health centers (rural and urban). Temperature curve readings were performed on site in the presence of the medical staff and results were compared to those recorded twice a day on conventional temperature charts using lamellar thermometers installed in refrigerators doors. Testo curves showed that liquid vaccine storage temperatures fell below freezing too frequently and that temperatures readings of door thermometers were often inaccurate. Testo readings also detected power outages in refrigeration units used in urban settings and flame extinctions in kerosene lamp refrigerators due to refrigerator breakdown or windy weather conditions before the rainy season. The main advantage of this monitoring method is to provide reliable data as a basis not only for detection of possible freezing of liquid vaccines but also for discussion of cold chain management and improvement with medical staff.
authors.affiliation: Agence de Médecine Preventive, Institut Pasteur, Paris, France. mschlumberger@wanadoo.fr
authors.forename: M
authors.initials: M
authors.lastname: Schlumberger
authors.forename: F
authors.initials: F
authors.lastname: Mireux
authors.forename: S G
authors.initials: SG
authors.lastname: Tchang
authors.forename: D
authors.initials: D
authors.lastname: Mboutbogol
authors.forename: D O
authors.initials: DO
authors.lastname: Cheikh
authors.forename: A A
authors.initials: AA
authors.lastname: Hissein
authors.forename: B O
authors.initials: BO
authors.lastname: Youssouf
authors.forename: M M
authors.initials: MM
authors.lastname: Brahimi
authors.forename: Y
authors.initials: Y
authors.lastname: Gamatié
mesh_terms: D002596:Chad; D003201:Computers; D004356:Drug Storage / Q000379:methods*; D006801:Humans; D012034:Refrigeration; D013821:Thermometers; D014329:Tropical Climate; D014612:Vaccines
chemical_list: D014612:Vaccines