title: [Research on procedure to assure the continued validity of the verification results of standard mercury-in-glass thermometer (grade I)].
abstract: Standard mercury-in-glass thermometer (Grade I) can be used to perform traceable measurements. Full recalibration is necessary to assure the continued validity of the verification results of the standard mercury-in-glass thermometer (Grade I). The present paper shows researches on procedure for the recalibration at the ice point of standard mercury-in-glass thermometer (Grade I), and points out different calculation ways of the true temperature of the thermostatic bath. The different values of scale correction and adjusted scale correction are compared in this paper.
authors.affiliation: Medical Engineering Support Center, Medical Support Division, Chinese PLA's General Hospital, Beijing 100853, China.
authors.forename: Juan
authors.initials: J
authors.lastname: Zhou
authors.forename: Guangrong
authors.initials: G
authors.lastname: Lin
authors.forename: Yong
authors.initials: Y
authors.lastname: Yan
authors.forename: Desen
authors.initials: D
authors.lastname: Cao
authors.forename: Hao
authors.initials: H
authors.lastname: Wu
mesh_terms: D005898:Glass; D008628:Mercury; D015203:Reproducibility of Results; D012680:Sensitivity and Specificity; D013696:Temperature; D013821:Thermometers / Q000592:standards*
chemical_list: D008628:Mercury