title: X-ray dose reduction through adaptive exposure in fluoroscopic imaging.
abstract: X-ray fluoroscopy is widely used for image guidance during cardiac intervention. However, radiation dose in these procedures can be high, and this is a significant concern, particularly in pediatric applications. Pediatrics procedures are in general much more complex than those performed on adults and thus are on average four to eight times longer. Furthermore, children can undergo up to 10 fluoroscopic procedures by the age of 10, and have been shown to have a three-fold higher risk of developing fatal cancer throughout their life than the general population. We have shown that radiation dose can be significantly reduced in adult cardiac procedures by using our scanning beam digital x-ray (SBDX) system-- a fluoroscopic imaging system that employs an inverse imaging geometry (Figure 1, Movie 1 and Figure 2). Instead of a single focal spot and an extended detector as used in conventional systems, our approach utilizes an extended X-ray source with multiple focal spots focused on a small detector. Our X-ray source consists of a scanning electron beam sequentially illuminating up to 9,000 focal spot positions. Each focal spot projects a small portion of the imaging volume onto the detector. In contrast to a conventional system where the final image is directly projected onto the detector, the SBDX uses a dedicated algorithm to reconstruct the final image from the 9,000 detector images. For pediatric applications, dose savings with the SBDX system are expected to be smaller than in adult procedures. However, the SBDX system allows for additional dose savings by implementing an electronic adaptive exposure technique. Key to this method is the multi-beam scanning technique of the SBDX system: rather than exposing every part of the image with the same radiation dose, we can dynamically vary the exposure depending on the opacity of the region exposed. Therefore, we can significantly reduce exposure in radiolucent areas and maintain exposure in more opaque regions. In our current implementation, the adaptive exposure requires user interaction (Figure 3). However, in the future, the adaptive exposure will be real time and fully automatic. We have performed experiments with an anthropomorphic phantom and compared measured radiation dose with and without adaptive exposure using a dose area product (DAP) meter. In the experiment presented here, we find a dose reduction of 30%.
authors.affiliation: Triple Ring Technologies.
authors.forename: Steve
authors.initials: S
authors.lastname: Burion
authors.forename: Tobias
authors.initials: T
authors.lastname: Funk
mesh_terms: D000328:Adult; D002648:Child; D005471:Fluoroscopy / Q000009:adverse effects / Q000295:instrumentation* / Q000379:methods*; D006801:Humans; D019047:Phantoms, Imaging; D011829:Radiation Dosage; D011832:Radiation Injuries / Q000517:prevention & control; D011856:Radiographic Image Enhancement / Q000379:methods; D011857:Radiographic Image Interpretation, Computer-Assisted / Q000379:methods*; D014965:X-Rays