"Detachment of the carinal hook following endobronchial intubation with a double lumen tube".
BACKGROUND
Carinal hooks increases difficulty at endotracheal intubation. Amputation of the carinal hook during passage and malpositioning of the tube to the hook are some of the potential problems related with left-sided Carlens double lumen tube (DLT). This article reports an amputation of the hook during a difficult selective intubation and aimed at calling the attention to complications associated with DLTs and the importance of fiberoptic bronchoscopy.


METHODS
A 68 year-old woman was scheduled for right-sided thoracotomy in whom blind DLT insertion was performed. Narrowed trachea causes difficulty in rotating the DLT 90° counter-clockwise. After carinal hook was noticed upon visual inspection of the DLT, fiberoptic bronchoscopy was used to remove the missing part (with the use of forceps) from the right mainstem bronchus.


CONCLUSIONS
Insertion of DLTs with carinal hook is associated with technical problems and potentially life-threatening hazards have discouraged their use. Fiberoptic evaluation and repositioning solves most of the problems. Although amputation of the carinal hook has not been previously reported, clinicians should be alert. This case report emphasizes the utility of the fiberoptic bronchoscopy in the operating theatre for placement, positioning and inspection of the carinal hook DLT.
Department of Anesthesiology, ULS de Castelo Branco, Portugal. carolinassrocha@gmail.com.
Ana C
AC
Rocha
Mafalda G
MG
Martins
Luísa I
LI
Silva
José M
JM
Nunes