Workplace aerosol mass concentration measurement using optical particle counters.
Direct-reading aerosol measurement usually uses the optical properties of airborne particles to detect and measure particle concentration. In the case of occupational hygiene, mass concentration measurement is often required. Two aerosol monitoring methods are based on the principle of light scattering: optical particle counting (OPC) and photometry. The former analyses the light scattered by a single particle, the latter by a cloud of particles. Both methods need calibration to transform the quantity of scattered light detected into particle concentration. Photometers are simpler to use and can be directly calibrated to measure mass concentration. However, their response varies not only with aerosol concentration but also with particle size distribution, which frequently contributes to biased measurement. Optical particle counters directly measure the particle number concentration and particle size that allows assessment of the particle mass provided the particles are spherical and of known density. An integrating algorithm is used to calculate the mass concentration of any conventional health-related aerosol fraction. The concentrations calculated thus have been compared with simultaneous measurements by conventional gravimetric sampling to check the possibility of field OPC calibration with real workplace aerosols with a view to further monitoring particle mass concentration. Aerosol concentrations were measured in the food industry using the OPC GRIMM® 1.108 and the CIP 10-Inhalable and CIP 10-Respirable (ARELCO®) aerosol samplers while meat sausages were being brushed and coated with calcium carbonate. Previously, the original OPC inlet had been adapted to sample inhalable aerosol. A mixed aerosol of calcium carbonate and fungi spores was present in the workplace. The OPC particle-size distribution and an estimated average particle density of both aerosol components were used to calculate the mass concentration. The inhalable and respirable aerosol fractions calculated from the OPC data are closely correlated with the results of the particle size-selective sampling using the CIP 10. Furthermore, the OPC data allow calculation of the thoracic fraction of workplace aerosol (not measured by sampling), which is interesting in the presence of allergenic particles like fungi spores. The results also show that the modified COP inlet adequately samples inhalable aerosol in the range of workplace particle-size distribution.
Institut National de Recherche et de Sécurité, 1 rue du Morvan, CS 60027, 54519 Vandoeuvre-lès-Nancy, France.
Peter
P
Görner
Xavier
X
Simon
Denis
D
Bémer
Göran
G
Lidén
D000336:Aerosols / Q000032:analysis*; D000395:Air Pollutants, Occupational / Q000032:analysis*; D016902:Air Pollution, Indoor / Q000032:analysis / Q000706:statistics & numerical data; D004784:Environmental Monitoring / Q000295:instrumentation / Q000379:methods*; D006801:Humans; D010316:Particle Size; D052638:Particulate Matter / Q000032:analysis; D010783:Photometry; D017132:Workplace / Q000706:statistics & numerical data
D000336:Aerosols; D000395:Air Pollutants, Occupational; D052638:Particulate Matter