title: Accuracy of Acuros XB and AAA dose calculation for small fields with reference to RapidArc(®) stereotactic treatments.
abstract: OBJECTIVE
  To assess the accuracy against measurements of two photon dose calculation algorithms (Acuros XB and the Anisotropic Analytical algorithm AAA) for small fields usable in stereotactic treatments with particular focus on RapidArc(®).
  
  
  METHODS
  Acuros XB and AAA were configured for stereotactic use. Baseline accuracy was assessed on small jaw-collimated open fields for different values for the spot sizes parameter in the beam data: 0.0, 0.5, 1, and 2 mm. Data were calculated with a grid of 1 × 1 mm(2). Investigated fields were: 3 × 3, 2 × 2, 1 × 1, and 0.8 × 0.8 cm(2) with a 6 MV photon beam generated from a Clinac2100iX (Varian, Palo Alto, CA). Profiles, PDD, and output factors were measured in water with a PTW diamond detector (detector size: 4 mm(2), thickness 0.4 mm) and compared to calculations. Four RapidArc test plans were optimized, calculated and delivered with jaw settings J3 × 3, J2 × 2, and J1 × 1 cm(2), the last was optimized twice to generate high (H) and low (L) modulation patterns. Each plan consisted of one partial arc (gantry 110° to 250°), and collimator 45°. Dose to isocenter was measured in a PTW Octavius phantom and compared to calculations. 2D measurements were performed by means of portal dosimetry with the GLAaS method developed at authors' institute. Analysis was performed with gamma pass-fail test with 3% dose difference and 2 mm distance to agreement thresholds.
  
  
  RESULTS
  Open square fields: penumbrae from open field profiles were in good agreement with diamond measurements for 1 mm spot size setting for Acuros XB, and between 0.5 and 1 mm for AAA. Maximum MU difference between calculations and measurements was 1.7% for Acuros XB (0.2% for fields greater than 1 × 1 cm(2)) with 0.5 or 1 mm spot size. Agreement for AAA was within 0.7% (2.8%) for 0.5 (1 mm) spot size. RapidArc plans: doses were evaluated in a 4 mm diameter structure at isocenter and computed values differed from measurements by 0.0, -0.2, 5.5, and -3.4% for Acuros XB calculations (1 mm spot size), and of -0.1, 0.3, 6.7, and -1.2% for AAA, respectively for J3 × 3, J2 × 2, J1 × 1H, J1 × 1L RapidArc plans. Gamma Agreement Index from 2D dose analysis was higher than 95% for J3 × 3 and J2 × 2 plans, being around 80% for J1 × 1 maps. Sensitivity with respect to the dosimetric leaf gap and transmission factor MLC parameters was evaluated in the four RapidArc plans, showing the need to properly set the dosimetric leaf gap for accurate calculations.
  
  
  CONCLUSIONS
  Acuros XB and AAA showed acceptable characteristics for stereotactic small fields if adequate tuning of configuration parameters is performed. Dose calculated for RapidArc stereotactic plans showed an acceptable agreement against point and 2D measurements. Both algorithms can therefore be considered safely applicable to stereotactic treatments.
authors: Oncology Institute of Southern Switzerland, Bellinzona, Switzerland. afc@iosi.ch
authors: Antonella
authors: A
authors: Fogliata
authors: Giorgia
authors: G
authors: Nicolini
authors: Alessandro
authors: A
authors: Clivio
authors: Eugenio
authors: E
authors: Vanetti
authors: Luca
authors: L
authors: Cozzi
mesh_terms: D000465:Algorithms; D017785:Photons / Q000627:therapeutic use; D011829:Radiation Dosage; D016634:Radiosurgery / Q000379:methods*; D011879:Radiotherapy Dosage; D011880:Radiotherapy Planning, Computer-Assisted / Q000379:methods*