title: A tunable general purpose Q-band resonator for CW and pulse EPR/ENDOR experiments with large sample access and optical excitation.
abstract: We describe a frequency tunable Q-band cavity (34 GHz) designed for CW and pulse Electron Paramagnetic Resonance (EPR) as well as Electron Nuclear Double Resonance (ENDOR) and Electron Electron Double Resonance (ELDOR) experiments. The TE(011) cylindrical resonator is machined either from brass or from graphite (which is subsequently gold plated), to improve the penetration of the 100 kHz field modulation signal. The (self-supporting) ENDOR coil consists of four 0.8mm silver posts at 2.67 mm distance from the cavity center axis, penetrating through the plunger heads. It is very robust and immune to mechanical vibrations. The coil is electrically shielded to enable CW ENDOR experiments with high RF power (500 W). The top plunger of the cavity is movable and allows a frequency tuning of ±2 GHz. In our setup the standard operation frequency is 34.0 GHz. The microwaves are coupled into the resonator through an iris in the cylinder wall and matching is accomplished by a sliding short in the coupling waveguide. Optical excitation of the sample is enabled through slits in the cavity wall (transmission ∼60%). The resonator accepts 3mm o.d. sample tubes. This leads to a favorable sensitivity especially for pulse EPR experiments of low concentration biological samples. The probehead dimensions are compatible with that of Bruker flexline Q-band resonators and it fits perfectly into an Oxford CF935 Helium flow cryostat (4-300 K). It is demonstrated that, due to the relatively large active sample volume (20-30 μl), the described resonator has superior concentration sensitivity as compared to commercial pulse Q-band resonators. The quality factor (Q(L)) of the resonator can be varied between 2600 (critical coupling) and 1300 (over-coupling). The shortest achieved π/2-pulse durations are 20 ns using a 3 W microwave amplifier. ENDOR (RF) π-pulses of 20 μs ((1)H @ 51 MHz) were obtained for a 300 W amplifier and 7 μs using a 2500 W amplifier. Selected applications of the resonator are presented.
authors.affiliation: Max-Planck-Institut für Bioanorganische Chemie, 45470 Mülheim an der Ruhr, Stiftstr. 34-36, Germany. reijerse@mpi-muelheim.mpg.de
authors.forename: Edward
authors.initials: E
authors.lastname: Reijerse
authors.forename: Friedhelm
authors.initials: F
authors.lastname: Lendzian
authors.forename: Roger
authors.initials: R
authors.lastname: Isaacson
authors.forename: Wolfgang
authors.initials: W
authors.lastname: Lubitz
mesh_terms: D004578:Electron Spin Resonance Spectroscopy / Q000295:instrumentation*; D004867:Equipment Design; D019544:Equipment Failure Analysis; D008027:Light; D008029:Lighting / Q000295:instrumentation*; D012815:Signal Processing, Computer-Assisted / Q000295:instrumentation*; D013048:Specimen Handling / Q000295:instrumentation*; D014159:Transducers