Measured and calculated K-fluorescence effects on the MTF of an amorphous-selenium based CCD x-ray detector.
OBJECTIVE
Theoretical reasoning suggests that direct conversion digital x-ray detectors based upon photoconductive amorphous-selenium (a-Se) could attain very high values of the MTF (modulation transfer function) at spatial frequencies well beyond 20 cycles mm(-1). One of the fundamental factors affecting resolution loss, particularly at x-ray energies just above the K-edge of selenium (12.66 keV), is the K-fluorescence reabsorption mechanism, wherein energy can be deposited in the detector at locations laterally displaced from the initial x-ray interaction site. This paper compares measured MTF changes above and below the Se K-edge of a CCD based a-Se x-ray detector with theoretical expectations.


METHODS
A prototype 25 μm sampling pitch (Nyquist frequency = 20 cycles mm(-1), 200 μm thick a-Se layer based x-ray detector, utilizing a specialized CCD readout device (200 × 400 area array), was used to make edge images with monochromatic x-rays above and below the K-edge of Se. A vacuum double crystal monochromator, exposed to polychromatic x-rays from a synchrotron, formed the monochromatic x-ray source. The monochromaticity of the x-rays was 99% or better. The presampling MTF was determined using the slanted edge method. The theory modeling the MTF performance of the detector includes the basic x-ray interaction physics in the a-Se layer as well as effects related to the operation of the CCD and charge trapping at a blocking layer present at the CCD/a-Se interface.


RESULTS
The MTF performance of the prototype a-Se CCD was reduced from the theoretical value prescribed by the basic Se x-ray interaction physics, principally by the presence of a blocking layer. Nevertheless, the K-fluorescence reduction in the MTF was observed, approximately as predicted by theory. For the CCD prototype detector, at five cycles mm(-1), there was a 14% reduction of the MTF, from a value of 0.7 below the K-edge of Se, to 0.6 just above the K-edge.


CONCLUSIONS
The MTF of an a-Se x-ray detector has been measured using monochromatic x-rays above and below the K-edge of selenium. The MTF is poorer above the K-edge by an amount consistent with theoretical expectations.
Sunnybrook Health Sciences Centre, Toronto, Ontario, Canada. david.hunter@sri.utoronto.ca
David M
DM
Hunter
George
G
Belev
Safa
S
Kasap
Martin J
MJ
Yaffe
D017076:Computer-Aided Design; D004867:Equipment Design; D019544:Equipment Failure Analysis; D005453:Fluorescence; D008962:Models, Theoretical; D012643:Selenium / Q000528:radiation effects*; D012815:Signal Processing, Computer-Assisted / Q000295:instrumentation*; D014057:Tomography, X-Ray Computed / Q000295:instrumentation*
D012643:Selenium