The first microbeam synchrotron X-ray fluorescence beamline at the Siam Photon Laboratory.
The first microbeam synchrotron X-ray fluorescence (µ-SXRF) beamline using continuous synchrotron radiation from Siam Photon Source has been constructed and commissioned as of August 2011. Utilizing an X-ray capillary half-lens allows synchrotron radiation from a 1.4 T bending magnet of the 1.2 GeV electron storage ring to be focused from a few millimeters-sized beam to a micrometer-sized beam. This beamline was originally designed for deep X-ray lithography (DXL) and was one of the first two operational beamlines at this facility. A modification has been carried out to the beamline in order to additionally enable µ-SXRF and synchrotron X-ray powder diffraction (SXPD). Modifications included the installation of a new chamber housing a Si(111) crystal to extract 8 keV synchrotron radiation from the white X-ray beam (for SXPD), a fixed aperture and three gate valves. Two end-stations incorporating optics and detectors for µ-SXRF and SXPD have then been installed immediately upstream of the DXL station, with the three techniques sharing available beam time. The µ-SXRF station utilizes a polycapillary half-lens for X-ray focusing. This optic focuses X-ray white beam from 5 mm × 2 mm (H × V) at the entrance of the lens down to a diameter of 100 µm FWHM measured at a sample position 22 mm (lens focal point) downstream of the lens exit. The end-station also incorporates an XYZ motorized sample holder with 25 mm travel per axis, a 5× ZEISS microscope objective with 5 mm × 5 mm field of view coupled to a CCD camera looking to the sample, and an AMPTEK single-element Si (PIN) solid-state detector for fluorescence detection. A graphic user interface data acquisition program using the LabVIEW platform has also been developed in-house to generate a series of single-column data which are compatible with available XRF data-processing software. Finally, to test the performance of the µ-SXRF beamline, an elemental surface profile has been obtained for a piece of ancient pottery from the Ban Chiang archaeological site, a UNESCO heritage site. It was found that the newly constructed µ-SXRF technique was able to clearly distinguish the distribution of different elements on the specimen.
Synchrotron Light Research Institute, 111 University Avenue, Muang District, Nakhon-Ratchasima 30000, Thailand. somchai@slri.or.th
Somchai
S
Tancharakorn
Waraporn
W
Tanthanuch
Nuntaporn
N
Kamonsutthipaijit
Narupon
N
Wongprachanukul
Methee
M
Sophon
Sarunyu
S
Chaichuay
Chunmanus
C
Uthaisar
Rattikorn
R
Yimnirun