A Monte Carlo simulation study of the effect of energy windows in computed tomography images based on an energy-resolved photon counting detector.
The energy-resolved photon counting detector provides the spectral information that can be used to generate images. The novel imaging methods, including the K-edge imaging, projection-based energy weighting imaging and image-based energy weighting imaging, are based on the energy-resolved photon counting detector and can be realized by using various energy windows or energy bins. The location and width of the energy windows or energy bins are important because these techniques generate an image using the spectral information defined by the energy windows or energy bins. In this study, the reconstructed images acquired with K-edge imaging, projection-based energy weighting imaging and image-based energy weighting imaging were simulated using the Monte Carlo simulation. The effect of energy windows or energy bins was investigated with respect to the contrast, coefficient-of-variation (COV) and contrast-to-noise ratio (CNR). The three images were compared with respect to the CNR. We modeled the x-ray computed tomography system based on the CdTe energy-resolved photon counting detector and polymethylmethacrylate phantom, which have iodine, gadolinium and blood. To acquire K-edge images, the lower energy thresholds were fixed at K-edge absorption energy of iodine and gadolinium and the energy window widths were increased from 1 to 25 bins. The energy weighting factors optimized for iodine, gadolinium and blood were calculated from 5, 10, 15, 19 and 33 energy bins. We assigned the calculated energy weighting factors to the images acquired at each energy bin. In K-edge images, the contrast and COV decreased, when the energy window width was increased. The CNR increased as a function of the energy window width and decreased above the specific energy window width. When the number of energy bins was increased from 5 to 15, the contrast increased in the projection-based energy weighting images. There is a little difference in the contrast, when the number of energy bin is increased from 15 to 33. The COV of the background in the projection-based energy weighting images is only slightly changed as a function of the number of energy bins. In the image-based energy weighting images, when the number of energy bins were increased, the contrast and COV increased and decreased, respectively. The CNR increased as a function of the number of energy bins. It was concluded that the image quality is dependent on the energy window, and an appropriate choice of the energy window is important to improve the image quality.
Department of Radiological Science, College of Health Science, Yonsei University, Wonju 220-710, Korea.
Seung-Wan
SW
Lee
Yu-Na
YN
Choi
Hyo-Min
HM
Cho
Young-Jin
YJ
Lee
Hyun-Ju
HJ
Ryu
Hee-Joung
HJ
Kim
D007091:Image Processing, Computer-Assisted; D009010:Monte Carlo Method; D017785:Photons; D014057:Tomography, X-Ray Computed / Q000379:methods*