NIST/ISAC standardization study: variability in assignment of intensity values to fluorescence standard beads and in cross calibration of standard beads to hard dyed beads.
Results from a standardization study cosponsored by the International Society for Advancement of Cytometry (ISAC) and the US National Institute of Standards and Technology (NIST) are reported. The study evaluated the variability of assigning intensity values to fluorophore standard beads by bead manufacturers and the variability of cross calibrating the standard beads to stained polymer beads (hard-dyed beads) using different flow cytometers. Hard dyed beads are generally not spectrally matched to the fluorophores used to stain cells, and spectral response varies among flow cytometers. Thus if hard dyed beads are used as fluorescence calibrators, one expects calibration for specific fluorophores (e.g., FITC or PE) to vary among different instruments. Using standard beads surface-stained with specific fluorophores (FITC, PE, APC, and Pacific Blue™), the study compared the measured intensity of fluorophore standard beads to that of hard dyed beads through cross calibration on 133 different flow cytometers. Using robust CV as a measure of variability, the variation of cross calibrated values was typically 20% or more for a particular hard dyed bead in a specific detection channel. The variation across different instrument models was often greater than the variation within a particular instrument model. As a separate part of the study, NIST and four bead manufacturers used a NIST supplied protocol and calibrated fluorophore solution standards to assign intensity values to the fluorophore beads. Values assigned to the reference beads by different groups varied by orders of magnitude in most cases, reflecting differences in instrumentation used to perform the calibration. The study concluded that the use of any spectrally unmatched hard dyed bead as a general fluorescence calibrator must be verified and characterized for every particular instrument model. Close interaction between bead manufacturers and NIST is recommended to have reliable and uniformly assigned fluorescence standard beads.
Biochemical Science Division, National Institute of Standards and Technology(NIST), Gaithersburg, Maryland 20899-8312, USA.
Robert A
RA
Hoffman
Lili
L
Wang
Martin
M
Bigos
John P
JP
Nolan
D002138:Calibration; D003374:Coumarins / Q000737:chemistry; D005434:Flow Cytometry / Q000295:instrumentation / Q000592:standards*; D019793:Fluorescein / Q000737:chemistry; D005456:Fluorescent Dyes / Q000737:chemistry; D008863:Microspheres; D010078:Oxazines / Q000737:chemistry; D012015:Reference Standards; D012016:Reference Values; D014481:United States; D037041:United States Government Agencies
D003374:Coumarins; D005456:Fluorescent Dyes; D010078:Oxazines; C044808:nile red; D019793:Fluorescein