title: Sensitivity maximized near-field scanning optical microscope with dithering sample stage.
abstract: We developed a new scheme for a higher sensitivity near-field scanning optical microscope (NSOM) by using a dithering sample stage rather than a dithering probe for the constant gap control between probe and sample. In a conventional NSOM, which use tip dithering feedback mechanism, the Q factor drastically decreases from 7783 to 1000 (13%) or even to 100 (1%) because harmonic oscillating characteristic is deteriorated owing to the large change of stiffness and mass of one prong of tuning fork when a probe is attached to it. In our proposed scheme, on the other hand, we use sample dithering feedback mechanism, where the probe is not attached to the tuning fork and the sample is loaded directly onto the surface of dithering tuning fork. Thus, the Q factor does not decrease significantly, from only 7783 to 7480 (96%), because the loaded sample hardly changes the stiffness and mass of tuning fork. Accordingly, gap control between the immobile fiber probe and the dithering sample is performed precisely by detecting the shear force with high sensitivity. Consequently, the extremely high Q factor enables clear observation of graphene sheets with sub-nanometer vertical resolution, which is not possible with a conventional NSOM setup.
authors.affiliation: Advanced Photonics Research Institute, Gwangju Institute of Science and Technology, Gwangju 500-712, South Korea.
authors.forename: Kyoung-Duck
authors.initials: KD
authors.lastname: Park
authors.forename: Seung Gol
authors.initials: SG
authors.lastname: Lee
authors.forename: Chaejeong
authors.initials: C
authors.lastname: Heo
authors.forename: Young Hee
authors.initials: YH
authors.lastname: Lee
authors.forename: Mun Seok
authors.initials: MS
authors.lastname: Jeong
mesh_terms: D006108:Graphite / Q000737:chemistry; D008853:Microscopy / Q000295:instrumentation*; D055096:Optical Devices; D013499:Surface Properties
chemical_list: D006108:Graphite