On the feasibility of treating to a 1.5 cm PTV with a commercial single-entry hybrid applicator in APBI breast brachytherapy.
OBJECTIVE
To evaluate and determine whether 30 patients previously treated with the SAVI™ device could have been treated to a PTV_EVAL created with a 1.5 cm expansion. This determination was based upon dosimetric parameters derived from current recommendations and dose-response data.


METHODS
Thirty patients were retrospectively planned with PTV_EVALs generated with a 1.5 cm expansion (PTV_EVAL_1.5). Plans were evaluated based on PTV_EVAL_1.5 coverage (V90, V95, V100), skin and rib maximum doses (0.1 cc maximum dose as a percentage of prescription dose), as well as V150 and V200 for the PTV_EVAL_1.5. The treatment planning goal was to deliver ≥90% of the prescribed dose to ≥90% of the PTV_EVAL_1.5. Skin and rib maximum doses were to be ≤125% of the prescription dose and preferably ≤100% of the prescription dose. V150 and V200 were not allowed to exceed 52.5 cc and 21 cc, respectively. Plans not meeting the above criteria were recomputed with a 1.25 cm expanded PTV_EVAL and re-evaluated.


RESULTS
Based on the above dose constraints, 30% (9/30) of the patients evaluated could have been treated with a 1.5 cm PTV_EVAL. The breakdown of cases successfully achieving the above dose constraints by applicator was: 0/4 (0%) 6-1, 6/15 (40%) 8-1, and 3/11 (27%) 10-1. For these PTV_EVAL_1.5 plans, median V90% was 90.3%, whereas the maximum skin and rib doses were all less than 115.2% and 117.6%, respectively. The median V150 and V200 volumes were 39.2 cc and 19.3, respectively. The treated PTV_EVAL_1.5 was greater in volume than the PTV_EVAL by 41.7 cc, and 60 cc for the 8-1, and 10-1 applicators, respectively. All remaining plans (17) successfully met the above dose constraints to be treated with a 1.25 cm PTV_EVAL (PTV_EVAL_1.25). For the PTV_EVAL_1.25 plans, V90% was 93.7%, and the maximum skin and rib doses were all less than 109.2% and 102.5%, respectively. The median V150 and V200 volumes were 41.2 cc and 19.3, respectively. The treated PTV_EVAL_1.25 was greater in volume than the PTV_EVAL by 16 cc, 24.9 cc, and 33.5 cc for the 6-1, 8-1 and 10-1 applicators, respectively.


CONCLUSIONS
It is dosimetrically possible to treat beyond the currently advised 1.0 cm expanded PTV_EVAL. Most patients should be able to be treated with a 1.25 cm PTV_EVAL and a select group with a 1.5 cm PTV_EVAL. Applicator size appears to determine the ability to expand to a 1.5 cm PTV_EVAL, as smaller devices were not as propitious in this regard. Further studies may identify additional patient groups that would benefit from this approach.
Department of Radiation Physics, The University of Texas M.D. Anderson Cancer Center.
Kent A
KA
Gifford
Christopher L
CL
Nelson
Steven M
SM
Kirsner
Kelly D
KD
Kisling
Matthew T
MT
Ballo
Elizabeth S
ES
Bloom
APBI; PTV_EVAL; SAVI; breast brachytherapy