title: [Methods of detector response function establishment in X-ray fluorescence spectra analysis].
abstract: During the measurement and analysis process of X-ray fluorescence spectra, it is very helpful to improve the analyze speed, accuracy and automaticity of X-ray fluorescence spectra analysis by establishing detector response function(DRF), which represents the shape of full energy peak and can provide former basic data for subsequent X-ray analysis technique. For the theory and model of semiconductor DRF in X-ray energy spectrum measurements, methods of three typical detector response function model establishment, key parameters of full energy peak standard deviation and Fano factor calculation, etc. are discussed, and meanwhile, the summarization and contrast of existing studies are shown in this paper. Finally, the suggestion for modeling methods of DRF in X-ray fluorescence spectra measurements is provided.
authors: State Key Laboratory of Geoharzard Prevention and Geoenvironment Protection, Chengdu University of Technology, Chengdu 610059, China. lizhe3017@163.com
authors: Zhe
authors: Z
authors: Li
authors: Xian-Guo
authors: XG
authors: Tuo
authors: Jian-Bo
authors: JB
authors: Yang
authors: Ming-Zhe
authors: MZ
authors: Liu
authors: Yi
authors: Y
authors: Cheng
authors: Lei
authors: L
authors: Wang
authors: Jian-Bin
authors: JB
authors: Zhou